-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 26 08:01:02 2022 UTC
-- File 20211002_003740_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.472367 to 35.362721
-each separately from 11s to 35s and from 362432us to 902693us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.429748 in 1 packets: 18
      12.459739 in 1 packets: 36
MIU packet time from 2021-10-02T00:37:54.472.367.000 to 2021-10-02T00:38:18.362.721.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.429748 in 1 packets: 18
      12.459739 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9091684 to 36.9090883
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999633 in 1 packets: 18
      12.9999566 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_003740_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3815367 to 13.3818257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.4297110  -- OK for SL --
   MIU time from 2021-10-02T00:37:54.472.367.000 to 2021-10-02T00:37:54.472.656.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.429748

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.8115367 to 24.8118257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4596950  -- OK for SL --
   MIU time from 2021-10-02T00:38:05.902.404.000 to 2021-10-02T00:38:05.902.693.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.459739

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2715207 to 37.2718097
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T00:38:18.362.432.000 to 2021-10-02T00:38:18.362.721.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.381536700 corrected by 40bit 10MHz counter of 12.909168400 obtained at 1Hz counter of 11. Result: 11.472369859
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T00:37:54.472.226.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T00:38:18.362.580.725
   Min 1st sample XO time among all SL packets: 11.472229583
   Max 1st sample XO time among all SL packets: 35.362582322
   TM values from -263 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T00:37:54.472.226.725 (XO: 11.472229583 - 11.472535584, t=37.2 degC)  Rank: 00079
   TM values: Max =   217  Min =  -263  Ave =    -4.45  Std =    19.93
   Min TM value:-263 at an offset of 118.725 usec -> 2021-10-02T00:37:54.472.345.450 (XO: 11.472348309)
   Max TM value:217 at an offset of 121.363 usec -> 2021-10-02T00:37:54.472.348.087 (XO: 11.472350946)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 116.300 usec -> 2021-10-02T00:37:54.472.343.025 (XO: 11.472345884)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 118.588 usec -> 2021-10-02T00:37:54.472.345.312 (XO: 11.472348171)

 - Block number: 2 starting at 2021-10-02T00:38:05.902.263.725 (XO: 22.902267702 - 22.902573703, t=37.2 degC)  Rank: 00090
   TM values: Max =   229  Min =  -253  Ave =    -4.39  Std =    28.60
   Min TM value:-253 at an offset of 117.913 usec -> 2021-10-02T00:38:05.902.381.637 (XO: 22.902385615)
   Max TM value:229 at an offset of 120.575 usec -> 2021-10-02T00:38:05.902.384.300 (XO: 22.902388278)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 120.438 usec -> 2021-10-02T00:38:05.902.384.162 (XO: 22.902388140)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 117.688 usec -> 2021-10-02T00:38:05.902.381.412 (XO: 22.902385390)

 - Block number: 3 starting at 2021-10-02T00:38:18.362.291.725 (XO: 35.362293321 - 35.362599322, t=37.2 degC)  Rank: 00079
   TM values: Max =   206  Min =  -256  Ave =    -4.42  Std =    20.17
   Min TM value:-256 at an offset of 109.575 usec -> 2021-10-02T00:38:18.362.401.300 (XO: 35.362402896)
   Max TM value:206 at an offset of 112.200 usec -> 2021-10-02T00:38:18.362.403.925 (XO: 35.362405521)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 112.000 usec -> 2021-10-02T00:38:18.362.403.725 (XO: 35.362405321)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 109.400 usec -> 2021-10-02T00:38:18.362.401.125 (XO: 35.362402721)
