-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 26 05:36:27 2022 UTC
-- File 20211001_225328_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.652850 to 25.003185
-each separately from 11s to 25s and from 2896us to 653139us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.349757 in 1 packets: 18
MIU packet time from 2021-10-01T22:53:42.652.850.000 to 2021-10-01T22:53:56.003.185.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.349757 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8236948 to 26.8236481
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_225328_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4765453 to 13.4768343
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3497110  -- OK for SL --
   MIU time from 2021-10-01T22:53:42.652.850.000 to 2021-10-01T22:53:42.653.139.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.349757

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.8265453 to 26.8268343
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T22:53:56.002.896.000 to 2021-10-01T22:53:56.003.185.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.476545300 corrected by 40bit 10MHz counter of 12.823694800 obtained at 1Hz counter of 11. Result: 11.652852720
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-01T22:53:42.652.709.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T22:53:56.003.044.725
   Min 1st sample XO time among all SL packets: 11.652712444
   Max 1st sample XO time among all SL packets: 25.003045935
   TM values from -256 to 231 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T22:53:42.652.709.725 (XO: 11.652712444 - 11.653018445, t=37.2 degC)  Rank: 00081
   TM values: Max =   220  Min =  -256  Ave =    -4.46  Std =    26.10
   Min TM value:-256 at an offset of 113.213 usec -> 2021-10-01T22:53:42.652.822.937 (XO: 11.652825657)
   Max TM value:220 at an offset of 128.975 usec -> 2021-10-01T22:53:42.652.838.700 (XO: 11.652841420)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 128.850 usec -> 2021-10-01T22:53:42.652.838.575 (XO: 11.652841295)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 113.038 usec -> 2021-10-01T22:53:42.652.822.762 (XO: 11.652825482)

 - Block number: 2 starting at 2021-10-01T22:53:56.002.755.725 (XO: 25.002756934 - 25.003062935, t=37.2 degC)  Rank: 00081
   TM values: Max =   231  Min =  -239  Ave =    -4.40  Std =    21.39
   Min TM value:-239 at an offset of 41.488 usec -> 2021-10-01T22:53:56.002.797.212 (XO: 25.002798422)
   Max TM value:231 at an offset of 44.163 usec -> 2021-10-01T22:53:56.002.799.887 (XO: 25.002801097)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 44.000 usec -> 2021-10-01T22:53:56.002.799.725 (XO: 25.002800935)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 41.300 usec -> 2021-10-01T22:53:56.002.797.025 (XO: 25.002798235)
