-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 23:48:07 2022 UTC
-- File 20211001_174110_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.696497 to 16.477060
-each separately from 6s to 16s and from 476771us to 696786us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.779985 in 1 packets: 18
MIU packet time from 2021-10-01T17:41:19.696.497.000 to 2021-10-01T17:41:29.477.060.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.779985 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8527342 to 17.8527007
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999665 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_174110_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5492322 to 8.5495212
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7799510  -- OK for SL --
   MIU time from 2021-10-01T17:41:19.696.497.000 to 2021-10-01T17:41:19.696.786.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.779985

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.3294722 to 18.3297612
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T17:41:29.476.771.000 to 2021-10-01T17:41:29.477.060.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.549232200 corrected by 40bit 10MHz counter of 7.852734200 obtained at 1Hz counter of 6. Result: 6.696500368
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-01T17:41:19.696.356.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T17:41:29.476.919.725
   Min 1st sample XO time among all SL packets: 6.696360093
   Max 1st sample XO time among all SL packets: 16.476921847
   TM values from -286 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T17:41:19.696.356.725 (XO: 6.696360093 - 6.696666094, t=37.2 degC)  Rank: 00079
   TM values: Max =   217  Min =  -281  Ave =    -4.51  Std =    26.68
   Min TM value:-281 at an offset of 251.063 usec -> 2021-10-01T17:41:19.696.607.788 (XO: 6.696611156)
   Max TM value:217 at an offset of 121.838 usec -> 2021-10-01T17:41:19.696.478.562 (XO: 6.696481931)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 251.188 usec -> 2021-10-01T17:41:19.696.607.913 (XO: 6.696611281)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 250.926 usec -> 2021-10-01T17:41:19.696.607.650 (XO: 6.696611018)

 - Block number: 2 starting at 2021-10-01T17:41:29.476.630.725 (XO: 16.476632846 - 16.476938847, t=37.2 degC)  Rank: 00090
   TM values: Max =   238  Min =  -286  Ave =    -4.46  Std =    24.53
   Min TM value:-286 at an offset of 109.088 usec -> 2021-10-01T17:41:29.476.739.812 (XO: 16.476741933)
   Max TM value:238 at an offset of 106.513 usec -> 2021-10-01T17:41:29.476.737.237 (XO: 16.476739358)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 109.200 usec -> 2021-10-01T17:41:29.476.739.925 (XO: 16.476742046)
   Min dTM/dt [1/us]:-2.77e+03 at an offset of 106.663 usec -> 2021-10-01T17:41:29.476.737.387 (XO: 16.476739508)
