-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 23:12:25 2022 UTC
-- File 20211001_170248_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.112224 to 18.862089
-each separately from 11s to 18s and from 112224us to 862089us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.749287 in 1 packets: 18
MIU packet time from 2021-10-01T17:03:02.112.224.000 to 2021-10-01T17:03:09.862.089.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.749287 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9192450 to 19.9192215
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999765 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_170248_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.0314706 to 13.0317596
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7492630  -- OK for SL --
   MIU time from 2021-10-01T17:03:02.112.224.000 to 2021-10-01T17:03:02.112.513.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.749287

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7810226 to 20.7813116
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T17:03:09.861.800.000 to 2021-10-01T17:03:09.862.089.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.031470600 corrected by 40bit 10MHz counter of 12.919245000 obtained at 1Hz counter of 11. Result: 11.112225970
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-01T17:03:02.112.083.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T17:03:09.861.948.725
   Min 1st sample XO time among all SL packets: 11.112085695
   Max 1st sample XO time among all SL packets: 18.861952669
   TM values from -262 to 268 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T17:03:02.112.083.725 (XO: 11.112085695 - 11.112391696, t=37.2 degC)  Rank: 00085
   TM values: Max =   249  Min =  -235  Ave =    -4.47  Std =    26.75
   Min TM value:-235 at an offset of 107.813 usec -> 2021-10-01T17:03:02.112.191.537 (XO: 11.112193508)
   Max TM value:249 at an offset of 110.463 usec -> 2021-10-01T17:03:02.112.194.187 (XO: 11.112196158)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 110.325 usec -> 2021-10-01T17:03:02.112.194.050 (XO: 11.112196020)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 107.600 usec -> 2021-10-01T17:03:02.112.191.325 (XO: 11.112193295)

 - Block number: 2 starting at 2021-10-01T17:03:09.861.659.725 (XO: 18.861663668 - 18.861969669, t=37.2 degC)  Rank: 00087
   TM values: Max =   268  Min =  -262  Ave =    -4.44  Std =    21.58
   Min TM value:-262 at an offset of 110.613 usec -> 2021-10-01T17:03:09.861.770.337 (XO: 18.861774281)
   Max TM value:268 at an offset of 113.138 usec -> 2021-10-01T17:03:09.861.772.862 (XO: 18.861776806)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 112.975 usec -> 2021-10-01T17:03:09.861.772.700 (XO: 18.861776644)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 110.438 usec -> 2021-10-01T17:03:09.861.770.162 (XO: 18.861774106)
