-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 13:11:30 2022 UTC
-- File 20211001_140614_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.779464 to 25.677733
-each separately from 12s to 25s and from 677444us to 779753us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.897691 in 1 packets: 18
MIU packet time from 2021-10-01T14:06:29.779.464.000 to 2021-10-01T14:06:42.677.733.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.897691 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8920878 to 26.8920439
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999561 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C1 -> 37.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_140614_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6715522 to 14.6718412
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8976470  -- OK for SL --
   MIU time from 2021-10-01T14:06:29.779.464.000 to 2021-10-01T14:06:29.779.753.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.897691

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.5694882 to 27.5697772
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T14:06:42.677.444.000 to 2021-10-01T14:06:42.677.733.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.671552200 corrected by 40bit 10MHz counter of 13.892087800 obtained at 1Hz counter of 12. Result: 12.779467050
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-01T14:06:29.779.323.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T14:06:42.677.592.725
   Min 1st sample XO time among all SL packets: 12.779326775
   Max 1st sample XO time among all SL packets: 25.677595329
   TM values from -258 to 249 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T14:06:29.779.323.725 (XO: 12.779326775 - 12.779632776, t=35.6 degC)  Rank: 00080
   TM values: Max =   244  Min =  -233  Ave =    -4.35  Std =    19.66
   Min TM value:-233 at an offset of 130.288 usec -> 2021-10-01T14:06:29.779.454.012 (XO: 12.779457063)
   Max TM value:244 at an offset of 133.038 usec -> 2021-10-01T14:06:29.779.456.762 (XO: 12.779459813)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 132.875 usec -> 2021-10-01T14:06:29.779.456.600 (XO: 12.779459650)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 118.350 usec -> 2021-10-01T14:06:29.779.442.075 (XO: 12.779445125)

 - Block number: 2 starting at 2021-10-01T14:06:42.677.303.725 (XO: 25.677306328 - 25.677612329, t=37.2 degC)  Rank: 00082
   TM values: Max =   249  Min =  -258  Ave =    -4.30  Std =    23.61
   Min TM value:-258 at an offset of 129.450 usec -> 2021-10-01T14:06:42.677.433.175 (XO: 25.677435778)
   Max TM value:249 at an offset of 132.100 usec -> 2021-10-01T14:06:42.677.435.825 (XO: 25.677438428)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 131.925 usec -> 2021-10-01T14:06:42.677.435.650 (XO: 25.677438253)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 129.213 usec -> 2021-10-01T14:06:42.677.432.937 (XO: 25.677435541)
