-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 08:22:12 2022 UTC
-- File 20211001_113752_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.967086 to 22.297274
-each separately from 9s to 22s and from 296985us to 967375us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329610 in 1 packets: 18
MIU packet time from 2021-10-01T11:38:04.967.086.000 to 2021-10-01T11:38:17.297.274.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329610 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8886754 to 23.8886319
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999565 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_113752_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8557620 to 11.8560510
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3295670  -- OK for SL --
   MIU time from 2021-10-01T11:38:04.967.086.000 to 2021-10-01T11:38:04.967.375.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.329610

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.1856180 to 24.1859070
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T11:38:17.296.985.000 to 2021-10-01T11:38:17.297.274.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.855762000 corrected by 40bit 10MHz counter of 10.888675400 obtained at 1Hz counter of 9. Result: 9.967089888
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-01T11:38:04.966.945.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T11:38:17.297.133.725
   Min 1st sample XO time among all SL packets: 9.966949613
   Max 1st sample XO time among all SL packets: 22.297135835
   TM values from -263 to 273 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T11:38:04.966.945.725 (XO: 9.966949613 - 9.967255614, t=37.2 degC)  Rank: 00086
   TM values: Max =   223  Min =  -263  Ave =    -4.40  Std =    25.57
   Min TM value:-263 at an offset of 105.263 usec -> 2021-10-01T11:38:04.967.050.987 (XO: 9.967054875)
   Max TM value:223 at an offset of 104.988 usec -> 2021-10-01T11:38:04.967.050.712 (XO: 9.967054600)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 107.688 usec -> 2021-10-01T11:38:04.967.053.412 (XO: 9.967057300)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 105.075 usec -> 2021-10-01T11:38:04.967.050.800 (XO: 9.967054688)

 - Block number: 2 starting at 2021-10-01T11:38:17.296.844.725 (XO: 22.296846834 - 22.297152835, t=37.2 degC)  Rank: 00086
   TM values: Max =   273  Min =  -261  Ave =    -4.38  Std =    22.60
   Min TM value:-261 at an offset of 106.200 usec -> 2021-10-01T11:38:17.296.950.925 (XO: 22.296953035)
   Max TM value:273 at an offset of 104.063 usec -> 2021-10-01T11:38:17.296.948.787 (XO: 22.296950897)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 103.950 usec -> 2021-10-01T11:38:17.296.948.675 (XO: 22.296950785)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 106.075 usec -> 2021-10-01T11:38:17.296.950.800 (XO: 22.296952910)
