-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 07:15:17 2022 UTC
-- File 20211001_110904_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.694050 to 25.391984
-each separately from 2s to 25s and from 391695us to 694339us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.697356 in 1 packets: 18
MIU packet time from 2021-10-01T11:09:09.694.050.000 to 2021-10-01T11:09:32.391.984.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.697356 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8904497 to 26.8903732
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999235 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_110904_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.5845008 to 4.5847898
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6972790  -- OK for SL --
   MIU time from 2021-10-01T11:09:09.694.050.000 to 2021-10-01T11:09:09.694.339.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.697356

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.2820688 to 27.2823578
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T11:09:32.391.695.000 to 2021-10-01T11:09:32.391.984.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.584500800 corrected by 40bit 10MHz counter of 3.890449700 obtained at 1Hz counter of 2. Result: 2.694053390
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-01T11:09:09.693.909.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T11:09:32.391.843.725
   Min 1st sample XO time among all SL packets: 2.693913115
   Max 1st sample XO time among all SL packets: 25.391845618
   TM values from -249 to 221 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T11:09:09.693.909.725 (XO: 2.693913115 - 2.694219116, t=37.2 degC)  Rank: 00087
   TM values: Max =   206  Min =  -231  Ave =    -4.43  Std =    19.99
   Min TM value:-231 at an offset of 97.563 usec -> 2021-10-01T11:09:09.694.007.287 (XO: 2.694010678)
   Max TM value:206 at an offset of 113.225 usec -> 2021-10-01T11:09:09.694.022.950 (XO: 2.694026340)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 113.063 usec -> 2021-10-01T11:09:09.694.022.787 (XO: 2.694026178)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 110.325 usec -> 2021-10-01T11:09:09.694.020.050 (XO: 2.694023440)

 - Block number: 2 starting at 2021-10-01T11:09:32.391.554.725 (XO: 25.391556617 - 25.391862618, t=37.2 degC)  Rank: 00083
   TM values: Max =   221  Min =  -249  Ave =    -4.43  Std =    23.10
   Min TM value:-249 at an offset of 125.200 usec -> 2021-10-01T11:09:32.391.679.925 (XO: 25.391681818)
   Max TM value:221 at an offset of 124.938 usec -> 2021-10-01T11:09:32.391.679.662 (XO: 25.391681555)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 127.638 usec -> 2021-10-01T11:09:32.391.682.362 (XO: 25.391684255)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 125.063 usec -> 2021-10-01T11:09:32.391.679.787 (XO: 25.391681680)
