-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 25 05:47:53 2022 UTC
-- File 20211001_100947_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.889738 to 74.569919
-each separately from 6s to 74s and from 149832us to 890027us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.729844 in 1 packets: 18
      16.529672 in 1 packets: 36
      12.319767 in 1 packets: 54
      23.099453 in 1 packets: 72
MIU packet time from 2021-10-01T10:09:56.889.738.000 to 2021-10-01T10:11:04.569.919.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.729844 in 1 packets: 18
      16.529672 in 1 packets: 36
      12.319767 in 1 packets: 54
      23.099453 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9242279 to 75.9239999
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999462 in 1 packets: 18
      16.9999430 in 1 packets: 36
      11.9999599 in 1 packets: 54
      22.9999229 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211001_100947_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8139663 to 8.8142553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7297910  -- OK for SL --
   MIU time from 2021-10-01T10:09:56.889.738.000 to 2021-10-01T10:09:56.890.027.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.729844

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5440463 to 24.5443353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5296150  -- OK for SL --
   MIU time from 2021-10-01T10:10:12.619.871.000 to 2021-10-01T10:10:12.620.160.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.529672

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.0739503 to 41.0742393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3197270  -- OK for SL --
   MIU time from 2021-10-01T10:10:29.149.832.000 to 2021-10-01T10:10:29.150.121.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.319767

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.3939663 to 53.3942553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.0993750  -- OK for SL --
   MIU time from 2021-10-01T10:10:41.469.888.000 to 2021-10-01T10:10:41.470.177.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.099453

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.4936303 to 76.4939193
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-01T10:11:04.569.630.000 to 2021-10-01T10:11:04.569.919.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.813966300 corrected by 40bit 10MHz counter of 7.924227900 obtained at 1Hz counter of 6. Result: 6.889741425
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-01T10:09:56.889.597.725
   Max 1st sample MIU time among all SL packets: 2021-10-01T10:11:04.569.778.725
   Min 1st sample XO time among all SL packets: 6.889601150
   Max 1st sample XO time among all SL packets: 74.569781005
   TM values from -244 to 200 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-01T10:09:56.889.597.725 (XO: 6.889601150 - 6.889907151, t=37.2 degC)  Rank: 00083
   TM values: Max =   200  Min =  -244  Ave =    -4.47  Std =    19.98
   Min TM value:-244 at an offset of 119.213 usec -> 2021-10-01T10:09:56.889.716.937 (XO: 6.889720363)
   Max TM value:200 at an offset of 121.738 usec -> 2021-10-01T10:09:56.889.719.462 (XO: 6.889722888)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 121.600 usec -> 2021-10-01T10:09:56.889.719.325 (XO: 6.889722750)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 119.038 usec -> 2021-10-01T10:09:56.889.716.762 (XO: 6.889720188)

 - Block number: 2 starting at 2021-10-01T10:10:12.619.730.725 (XO: 22.619733970 - 22.620039971, t=37.2 degC)  Rank: 00074
   TM values: Max =   185  Min =  -233  Ave =    -4.45  Std =    19.73
   Min TM value:-233 at an offset of 107.250 usec -> 2021-10-01T10:10:12.619.837.975 (XO: 22.619841220)
   Max TM value:185 at an offset of 106.975 usec -> 2021-10-01T10:10:12.619.837.700 (XO: 22.619840945)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 109.675 usec -> 2021-10-01T10:10:12.619.840.400 (XO: 22.619843645)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 107.088 usec -> 2021-10-01T10:10:12.619.837.812 (XO: 22.619841058)

 - Block number: 3 starting at 2021-10-01T10:10:29.149.691.725 (XO: 39.149693419 - 39.149999420, t=37.2 degC)  Rank: 00074
   TM values: Max =   195  Min =  -229  Ave =    -4.52  Std =    17.95
   Min TM value:-229 at an offset of 116.450 usec -> 2021-10-01T10:10:29.149.808.175 (XO: 39.149809869)
   Max TM value:195 at an offset of 116.125 usec -> 2021-10-01T10:10:29.149.807.850 (XO: 39.149809544)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 118.813 usec -> 2021-10-01T10:10:29.149.810.537 (XO: 39.149812232)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 116.250 usec -> 2021-10-01T10:10:29.149.807.975 (XO: 39.149809669)

 - Block number: 4 starting at 2021-10-01T10:10:41.469.747.725 (XO: 51.469750575 - 51.470056576, t=37.2 degC)  Rank: 00079
   TM values: Max =   189  Min =  -230  Ave =    -4.52  Std =    18.48
   Min TM value:-230 at an offset of 105.163 usec -> 2021-10-01T10:10:41.469.852.887 (XO: 51.469855738)
   Max TM value:189 at an offset of 107.788 usec -> 2021-10-01T10:10:41.469.855.512 (XO: 51.469858363)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 107.588 usec -> 2021-10-01T10:10:41.469.855.312 (XO: 51.469858163)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 104.938 usec -> 2021-10-01T10:10:41.469.852.662 (XO: 51.469855513)

 - Block number: 5 starting at 2021-10-01T10:11:04.569.489.725 (XO: 74.569492004 - 74.569798005, t=37.2 degC)  Rank: 00084
   TM values: Max =   199  Min =  -244  Ave =    -4.45  Std =    19.24
   Min TM value:-244 at an offset of 117.438 usec -> 2021-10-01T10:11:04.569.607.162 (XO: 74.569609442)
   Max TM value:199 at an offset of 120.000 usec -> 2021-10-01T10:11:04.569.609.725 (XO: 74.569612005)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 119.813 usec -> 2021-10-01T10:11:04.569.609.537 (XO: 74.569611817)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 117.288 usec -> 2021-10-01T10:11:04.569.607.012 (XO: 74.569609292)
