-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 04:04:17 2022 UTC
-- File 20210926_101212_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.794975 to 18.573749
-each separately from 7s to 18s and from 573460us to 795264us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.778196 in 1 packets: 18
MIU packet time from 2021-09-26T10:12:22.794.975.000 to 2021-09-26T10:12:33.573.749.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.778196 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9139413 to 19.9139047
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999634 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210926_101212_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7089176 to 9.7092066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7781590  -- OK for SL --
   MIU time from 2021-09-26T10:12:22.794.975.000 to 2021-09-26T10:12:22.795.264.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.778196

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      155 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.4873656 to 20.4876546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-26T10:12:33.573.460.000 to 2021-09-26T10:12:33.573.749.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.708917600 corrected by 40bit 10MHz counter of 8.913941300 obtained at 1Hz counter of 7. Result: 7.794978923
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-26T10:12:22.794.834.725
   Max 1st sample MIU time among all SL packets: 2021-09-26T10:12:33.573.608.725
   Min 1st sample XO time among all SL packets: 7.794838648
   Max 1st sample XO time among all SL packets: 18.573611575
   TM values from -504 to 438 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-26T10:12:22.794.834.725 (XO: 7.794838648 - 7.795144649, t=37.2 degC)  Rank: 00080
   TM values: Max =   438  Min =  -143  Ave =    -4.46  Std =    25.84
   Min TM value:-143 at an offset of 132.538 usec -> 2021-09-26T10:12:22.794.967.262 (XO: 7.794971186)
   Max TM value:438 at an offset of 147.363 usec -> 2021-09-26T10:12:22.794.982.087 (XO: 7.794986011)
   Max dTM/dt [1/us]:2.65e+03 at an offset of 147.313 usec -> 2021-09-26T10:12:22.794.982.037 (XO: 7.794985961)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 147.413 usec -> 2021-09-26T10:12:22.794.982.137 (XO: 7.794986061)

 - Block number: 2 starting at 2021-09-26T10:12:33.573.319.725 (XO: 18.573322574 - 18.573628575, t=37.2 degC)  Rank: 00155
   TM values: Max =   337  Min =  -504  Ave =    -4.16  Std =    59.30
   Min TM value:-504 at an offset of 146.013 usec -> 2021-09-26T10:12:33.573.465.737 (XO: 18.573468587)
   Max TM value:337 at an offset of 145.700 usec -> 2021-09-26T10:12:33.573.465.425 (XO: 18.573468275)
   Max dTM/dt [1/us]:4.00e+03 at an offset of 133.463 usec -> 2021-09-26T10:12:33.573.453.187 (XO: 18.573456037)
   Min dTM/dt [1/us]:-3.97e+03 at an offset of 113.363 usec -> 2021-09-26T10:12:33.573.433.087 (XO: 18.573435937)
