-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 01:32:09 2022 UTC
-- File 20210926_063746_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.280695 to 18.139596
-each separately from 12s to 18s and from 139307us to 280984us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.858323 in 1 packets: 18
MIU packet time from 2021-09-26T06:38:01.280.695.000 to 2021-09-26T06:38:07.139.596.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.858323 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9112489 to 19.9112287
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999798 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 3 packets: 19, 20, 21
      2 in 15 packets: 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210926_063746_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.1919450 to 14.1922340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.8583030  -- OK for SL --
   MIU time from 2021-09-26T06:38:01.280.695.000 to 2021-09-26T06:38:01.280.984.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.858323

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      125 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0505370 to 20.0508260
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-26T06:38:07.139.307.000 to 2021-09-26T06:38:07.139.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.191945000 corrected by 40bit 10MHz counter of 13.911248900 obtained at 1Hz counter of 12. Result: 12.280697054
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-09-26T06:38:01.280.554.725
   Max 1st sample MIU time among all SL packets: 2021-09-26T06:38:07.139.455.725
   Min 1st sample XO time among all SL packets: 12.280556779
   Max 1st sample XO time among all SL packets: 18.139457499
   TM values from -387 to 328 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-26T06:38:01.280.554.725 (XO: 12.280556779 - 12.280862780, t=37.2 degC)  Rank: 00068
   TM values: Max =   247  Min =  -221  Ave =    -4.52  Std =    27.23
   Min TM value:-221 at an offset of 85.050 usec -> 2021-09-26T06:38:01.280.639.775 (XO: 12.280641829)
   Max TM value:247 at an offset of 84.513 usec -> 2021-09-26T06:38:01.280.639.237 (XO: 12.280641292)
   Max dTM/dt [1/us]:1.19e+03 at an offset of 84.375 usec -> 2021-09-26T06:38:01.280.639.100 (XO: 12.280641154)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 84.913 usec -> 2021-09-26T06:38:01.280.639.637 (XO: 12.280641692)

 - Block number: 2 starting at 2021-09-26T06:38:07.139.166.725 (XO: 18.139168498 - 18.139474499, t=37.2 degC)  Rank: 00125
   TM values: Max =   328  Min =  -387  Ave =    -4.45  Std =    54.73
   Min TM value:-387 at an offset of 43.200 usec -> 2021-09-26T06:38:07.139.209.925 (XO: 18.139211698)
   Max TM value:328 at an offset of 77.938 usec -> 2021-09-26T06:38:07.139.244.662 (XO: 18.139246436)
   Max dTM/dt [1/us]:3.05e+03 at an offset of 43.250 usec -> 2021-09-26T06:38:07.139.209.975 (XO: 18.139211748)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 90.875 usec -> 2021-09-26T06:38:07.139.257.600 (XO: 18.139259373)
