-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 14:55:37 2022 UTC
-- File 20210924_094049_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.935495 to 16.635270
-each separately from 2s to 16s and from 634981us to 935784us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.699197 in 1 packets: 18
MIU packet time from 2021-09-24T09:40:54.935.495.000 to 2021-09-24T09:41:08.635.270.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.699197 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9269869 to 17.9269405
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999536 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210924_094049_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8624828 to 4.8627718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6991510  -- OK for SL --
   MIU time from 2021-09-24T09:40:54.935.495.000 to 2021-09-24T09:40:54.935.784.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.699197

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5619228 to 18.5622118
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-24T09:41:08.634.981.000 to 2021-09-24T09:41:08.635.270.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.862482800 corrected by 40bit 10MHz counter of 3.926986900 obtained at 1Hz counter of 2. Result: 2.935498987
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-24T09:40:54.935.354.725
   Max 1st sample MIU time among all SL packets: 2021-09-24T09:41:08.635.129.725
   Min 1st sample XO time among all SL packets: 2.935358712
   Max 1st sample XO time among all SL packets: 16.635133121
   TM values from -244 to 201 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-24T09:40:54.935.354.725 (XO: 2.935358712 - 2.935664713, t=37.2 degC)  Rank: 00076
   TM values: Max =   200  Min =  -237  Ave =    -4.52  Std =    16.83
   Min TM value:-237 at an offset of 125.325 usec -> 2021-09-24T09:40:54.935.480.050 (XO: 2.935484037)
   Max TM value:200 at an offset of 127.850 usec -> 2021-09-24T09:40:54.935.482.575 (XO: 2.935486562)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 127.700 usec -> 2021-09-24T09:40:54.935.482.425 (XO: 2.935486412)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 125.150 usec -> 2021-09-24T09:40:54.935.479.875 (XO: 2.935483862)

 - Block number: 2 starting at 2021-09-24T09:41:08.634.840.725 (XO: 16.634844120 - 16.635150121, t=37.2 degC)  Rank: 00082
   TM values: Max =   201  Min =  -244  Ave =    -4.48  Std =    20.49
   Min TM value:-244 at an offset of 122.250 usec -> 2021-09-24T09:41:08.634.962.975 (XO: 16.634966370)
   Max TM value:201 at an offset of 124.775 usec -> 2021-09-24T09:41:08.634.965.500 (XO: 16.634968895)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 124.625 usec -> 2021-09-24T09:41:08.634.965.350 (XO: 16.634968745)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 122.075 usec -> 2021-09-24T09:41:08.634.962.800 (XO: 16.634966195)
