-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 14:48:03 2022 UTC
-- File 20210924_093651_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.445814 to 44.204497
-each separately from 12s to 44s and from 55000us to 446103us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.608897 in 1 packets: 18
      17.148919 in 1 packets: 36
MIU packet time from 2021-09-24T09:37:06.445.814.000 to 2021-09-24T09:37:38.204.497.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.608897 in 1 packets: 18
      17.148919 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9346139 to 45.9345079
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999503 in 1 packets: 18
      16.9999437 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210924_093651_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.3804288 to 14.3807178
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6088470  -- OK for SL --
   MIU time from 2021-09-24T09:37:06.445.814.000 to 2021-09-24T09:37:06.446.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.608897

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.9895648 to 28.9898538
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1488630  -- OK for SL --
   MIU time from 2021-09-24T09:37:21.055.000.000 to 2021-09-24T09:37:21.055.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.148919

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.1387168 to 46.1390058
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-24T09:37:38.204.208.000 to 2021-09-24T09:37:38.204.497.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.380428800 corrected by 40bit 10MHz counter of 13.934613900 obtained at 1Hz counter of 12. Result: 12.445816416
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-24T09:37:06.445.673.725
   Max 1st sample MIU time among all SL packets: 2021-09-24T09:37:38.204.356.725
   Min 1st sample XO time among all SL packets: 12.445676140
   Max 1st sample XO time among all SL packets: 44.204358279
   TM values from -246 to 195 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-24T09:37:06.445.673.725 (XO: 12.445676140 - 12.445982141, t=37.2 degC)  Rank: 00079
   TM values: Max =   195  Min =  -246  Ave =    -4.54  Std =    28.01
   Min TM value:-246 at an offset of 191.826 usec -> 2021-09-24T09:37:06.445.865.550 (XO: 12.445867966)
   Max TM value:195 at an offset of 121.538 usec -> 2021-09-24T09:37:06.445.795.262 (XO: 12.445797678)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 124.163 usec -> 2021-09-24T09:37:06.445.797.887 (XO: 12.445800303)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 121.650 usec -> 2021-09-24T09:37:06.445.795.375 (XO: 12.445797791)

 - Block number: 2 starting at 2021-09-24T09:37:21.054.859.725 (XO: 27.054860512 - 27.055166513, t=37.2 degC)  Rank: 00076
   TM values: Max =   172  Min =  -228  Ave =    -4.43  Std =    18.22
   Min TM value:-228 at an offset of 106.375 usec -> 2021-09-24T09:37:21.054.966.100 (XO: 27.054966887)
   Max TM value:172 at an offset of 108.963 usec -> 2021-09-24T09:37:21.054.968.687 (XO: 27.054969474)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 108.813 usec -> 2021-09-24T09:37:21.054.968.537 (XO: 27.054969324)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 106.225 usec -> 2021-09-24T09:37:21.054.965.950 (XO: 27.054966737)

 - Block number: 3 starting at 2021-09-24T09:37:38.204.067.725 (XO: 44.204069278 - 44.204375279, t=37.2 degC)  Rank: 00082
   TM values: Max =   185  Min =  -228  Ave =    -4.55  Std =    19.44
   Min TM value:-228 at an offset of 105.750 usec -> 2021-09-24T09:37:38.204.173.475 (XO: 44.204175028)
   Max TM value:185 at an offset of 108.325 usec -> 2021-09-24T09:37:38.204.176.050 (XO: 44.204177603)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 108.138 usec -> 2021-09-24T09:37:38.204.175.862 (XO: 44.204177416)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 105.550 usec -> 2021-09-24T09:37:38.204.173.275 (XO: 44.204174828)
