-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 08:42:47 2022 UTC
-- File 20210923_221606_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.584468 to 19.245088
-each separately from 6s to 19s and from 244799us to 584757us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.660042 in 1 packets: 18
MIU packet time from 2021-09-23T22:16:15.584.468.000 to 2021-09-23T22:16:28.245.088.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.660042 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8706182 to 20.8705750
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999568 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210923_221606_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.4550873 to 8.4553763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6599990  -- OK for SL --
   MIU time from 2021-09-23T22:16:15.584.468.000 to 2021-09-23T22:16:15.584.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.660042

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1153753 to 21.1156643
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-23T22:16:28.244.799.000 to 2021-09-23T22:16:28.245.088.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.455087300 corrected by 40bit 10MHz counter of 7.870618200 obtained at 1Hz counter of 6. Result: 6.584471029
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-23T22:16:15.584.327.725
   Max 1st sample MIU time among all SL packets: 2021-09-23T22:16:28.244.947.725
   Min 1st sample XO time among all SL packets: 6.584330753
   Max 1st sample XO time among all SL packets: 19.244949833
   TM values from -246 to 226 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-23T22:16:15.584.327.725 (XO: 6.584330753 - 6.584636754, t=37.2 degC)  Rank: 00084
   TM values: Max =   226  Min =  -246  Ave =    -4.57  Std =    22.78
   Min TM value:-246 at an offset of 83.075 usec -> 2021-09-23T22:16:15.584.410.800 (XO: 6.584413829)
   Max TM value:226 at an offset of 113.925 usec -> 2021-09-23T22:16:15.584.441.650 (XO: 6.584444679)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 113.763 usec -> 2021-09-23T22:16:15.584.441.487 (XO: 6.584444516)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 82.925 usec -> 2021-09-23T22:16:15.584.410.650 (XO: 6.584413679)

 - Block number: 2 starting at 2021-09-23T22:16:28.244.658.725 (XO: 19.244660832 - 19.244966833, t=37.2 degC)  Rank: 00083
   TM values: Max =   211  Min =  -243  Ave =    -4.50  Std =    21.26
   Min TM value:-243 at an offset of 123.363 usec -> 2021-09-23T22:16:28.244.782.087 (XO: 19.244784195)
   Max TM value:211 at an offset of 123.088 usec -> 2021-09-23T22:16:28.244.781.812 (XO: 19.244783920)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 125.763 usec -> 2021-09-23T22:16:28.244.784.487 (XO: 19.244786595)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 123.200 usec -> 2021-09-23T22:16:28.244.781.925 (XO: 19.244784033)
