-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 06:41:54 2022 UTC
-- File 20210923_201504_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.269525 to 28.519770
-each separately from 3s to 28s and from 269525us to 519770us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.249667 in 1 packets: 18
MIU packet time from 2021-09-23T20:15:10.269.525.000 to 2021-09-23T20:15:35.519.770.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.249667 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7893115 to 29.7892282
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999167 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210923_201504_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.0588379 to 5.0591269
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2495830  -- OK for SL --
   MIU time from 2021-09-23T20:15:10.269.525.000 to 2021-09-23T20:15:10.269.814.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.249667

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.3087099 to 30.3089989
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-23T20:15:35.519.481.000 to 2021-09-23T20:15:35.519.770.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.058837900 corrected by 40bit 10MHz counter of 4.789311500 obtained at 1Hz counter of 3. Result: 3.269527289
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-23T20:15:10.269.384.725
   Max 1st sample MIU time among all SL packets: 2021-09-23T20:15:35.519.629.725
   Min 1st sample XO time among all SL packets: 3.269387014
   Max 1st sample XO time among all SL packets: 28.519632192
   TM values from -273 to 247 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-23T20:15:10.269.384.725 (XO: 3.269387014 - 3.269693015, t=37.2 degC)  Rank: 00092
   TM values: Max =   207  Min =  -273  Ave =    -4.52  Std =    30.08
   Min TM value:-273 at an offset of 166.213 usec -> 2021-09-23T20:15:10.269.550.938 (XO: 3.269553227)
   Max TM value:207 at an offset of 165.938 usec -> 2021-09-23T20:15:10.269.550.663 (XO: 3.269552952)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 163.738 usec -> 2021-09-23T20:15:10.269.548.463 (XO: 3.269550752)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 139.275 usec -> 2021-09-23T20:15:10.269.524.000 (XO: 3.269526289)

 - Block number: 2 starting at 2021-09-23T20:15:35.519.340.725 (XO: 28.519343191 - 28.519649192, t=37.2 degC)  Rank: 00085
   TM values: Max =   247  Min =  -268  Ave =    -4.59  Std =    24.79
   Min TM value:-268 at an offset of 112.963 usec -> 2021-09-23T20:15:35.519.453.687 (XO: 28.519456154)
   Max TM value:247 at an offset of 113.225 usec -> 2021-09-23T20:15:35.519.453.950 (XO: 28.519456416)
   Max dTM/dt [1/us]:2.61e+03 at an offset of 113.088 usec -> 2021-09-23T20:15:35.519.453.812 (XO: 28.519456279)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 110.575 usec -> 2021-09-23T20:15:35.519.451.300 (XO: 28.519453766)
