-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 06:05:50 2022 UTC
-- File 20210923_191141_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.417683 to 50.017560
-each separately from 1s to 50s and from 17271us to 577695us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.999506 in 1 packets: 18
      12.159639 in 1 packets: 36
      16.439576 in 1 packets: 54
MIU packet time from 2021-09-23T19:11:45.417.683.000 to 2021-09-23T19:12:34.017.560.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.999506 in 1 packets: 18
      12.159639 in 1 packets: 36
      16.439576 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9130055 to 51.9128421
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999333 in 1 packets: 18
      11.9999600 in 1 packets: 36
      16.9999433 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210923_191141_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3306898 to 3.3309788
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9994390  -- OK for SL --
   MIU time from 2021-09-23T19:11:45.417.683.000 to 2021-09-23T19:11:45.417.972.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.999506

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.3304178 to 23.3307068
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1595990  -- OK for SL --
   MIU time from 2021-09-23T19:12:05.417.478.000 to 2021-09-23T19:12:05.417.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.159639

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4903058 to 35.4905948
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4395190  -- OK for SL --
   MIU time from 2021-09-23T19:12:17.577.406.000 to 2021-09-23T19:12:17.577.695.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.439576

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.9301138 to 51.9304028
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-23T19:12:34.017.271.000 to 2021-09-23T19:12:34.017.560.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.330689800 corrected by 40bit 10MHz counter of 2.913005500 obtained at 1Hz counter of 1. Result: 1.417685678
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-23T19:11:45.417.542.725
   Max 1st sample MIU time among all SL packets: 2021-09-23T19:12:34.017.419.725
   Min 1st sample XO time among all SL packets: 1.417545403
   Max 1st sample XO time among all SL packets: 50.017420484
   TM values from -278 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-23T19:11:45.417.542.725 (XO: 1.417545403 - 1.417851404, t=37.2 degC)  Rank: 00079
   TM values: Max =   243  Min =  -256  Ave =    -4.48  Std =    24.59
   Min TM value:-256 at an offset of 126.388 usec -> 2021-09-23T19:11:45.417.669.112 (XO: 1.417671791)
   Max TM value:243 at an offset of 129.100 usec -> 2021-09-23T19:11:45.417.671.825 (XO: 1.417674503)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 128.875 usec -> 2021-09-23T19:11:45.417.671.600 (XO: 1.417674278)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 113.800 usec -> 2021-09-23T19:11:45.417.656.525 (XO: 1.417659203)

 - Block number: 2 starting at 2021-09-23T19:12:05.417.337.725 (XO: 21.417340144 - 21.417646145, t=37.2 degC)  Rank: 00076
   TM values: Max =   226  Min =  -278  Ave =    -4.44  Std =    25.70
   Min TM value:-278 at an offset of 112.975 usec -> 2021-09-23T19:12:05.417.450.700 (XO: 21.417453119)
   Max TM value:226 at an offset of 115.575 usec -> 2021-09-23T19:12:05.417.453.300 (XO: 21.417455719)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 115.375 usec -> 2021-09-23T19:12:05.417.453.100 (XO: 21.417455519)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 112.825 usec -> 2021-09-23T19:12:05.417.450.550 (XO: 21.417452969)

 - Block number: 3 starting at 2021-09-23T19:12:17.577.265.725 (XO: 33.577268688 - 33.577574689, t=37.2 degC)  Rank: 00082
   TM values: Max =   248  Min =  -265  Ave =    -4.57  Std =    27.60
   Min TM value:-265 at an offset of 114.188 usec -> 2021-09-23T19:12:17.577.379.912 (XO: 33.577382876)
   Max TM value:248 at an offset of 116.800 usec -> 2021-09-23T19:12:17.577.382.525 (XO: 33.577385488)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 116.575 usec -> 2021-09-23T19:12:17.577.382.300 (XO: 33.577385263)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 114.038 usec -> 2021-09-23T19:12:17.577.379.762 (XO: 33.577382726)

 - Block number: 4 starting at 2021-09-23T19:12:34.017.130.725 (XO: 50.017131483 - 50.017437484, t=37.2 degC)  Rank: 00074
   TM values: Max =   242  Min =  -254  Ave =    -4.49  Std =    29.96
   Min TM value:-254 at an offset of 101.775 usec -> 2021-09-23T19:12:34.017.232.500 (XO: 50.017233259)
   Max TM value:242 at an offset of 116.513 usec -> 2021-09-23T19:12:34.017.247.237 (XO: 50.017247996)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 116.375 usec -> 2021-09-23T19:12:34.017.247.100 (XO: 50.017247859)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 113.638 usec -> 2021-09-23T19:12:34.017.244.362 (XO: 50.017245121)
