-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 04:49:45 2022 UTC
-- File 20210923_173811_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.729367 to 54.650526
-each separately from 14s to 54s and from 269873us to 909692us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.179747 in 1 packets: 18
      20.360181 in 1 packets: 36
      18.380075 in 1 packets: 54
MIU packet time from 2021-09-23T17:38:28.729.367.000 to 2021-09-23T17:39:08.650.526.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.179747 in 1 packets: 18
      20.360181 in 1 packets: 36
      18.380075 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8100655 to 55.8099321
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.9999967 in 1 packets: 18
      20.9999299 in 1 packets: 36
      17.9999400 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210923_173811_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5394337 to 16.5397227
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.1797430  -- OK for SL --
   MIU time from 2021-09-23T17:38:28.729.367.000 to 2021-09-23T17:38:28.729.656.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.179747

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.7194657 to 17.7197547
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3601110  -- OK for SL --
   MIU time from 2021-09-23T17:38:29.909.403.000 to 2021-09-23T17:38:29.909.692.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.360181

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.0798657 to 38.0801547
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3800150  -- OK for SL --
   MIU time from 2021-09-23T17:38:50.269.873.000 to 2021-09-23T17:38:50.270.162.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.380075

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.4601697 to 56.4604587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-23T17:39:08.650.237.000 to 2021-09-23T17:39:08.650.526.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.539433700 corrected by 40bit 10MHz counter of 15.810065500 obtained at 1Hz counter of 14. Result: 14.729370607
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-23T17:38:28.729.226.725
   Max 1st sample MIU time among all SL packets: 2021-09-23T17:39:08.650.385.725
   Min 1st sample XO time among all SL packets: 14.729230331
   Max 1st sample XO time among all SL packets: 54.650388471
   TM values from -264 to 264 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-23T17:38:28.729.226.725 (XO: 14.729230331 - 14.729536332, t=37.2 degC)  Rank: 00078
   TM values: Max =   239  Min =  -246  Ave =    -4.43  Std =    22.26
   Min TM value:-246 at an offset of 117.838 usec -> 2021-09-23T17:38:28.729.344.562 (XO: 14.729348169)
   Max TM value:239 at an offset of 120.425 usec -> 2021-09-23T17:38:28.729.347.150 (XO: 14.729350757)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 120.275 usec -> 2021-09-23T17:38:28.729.347.000 (XO: 14.729350607)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 117.625 usec -> 2021-09-23T17:38:28.729.344.350 (XO: 14.729347957)

 - Block number: 2 starting at 2021-09-23T17:38:29.909.262.725 (XO: 15.909266317 - 15.909572318, t=37.2 degC)  Rank: 00078
   TM values: Max =   248  Min =  -264  Ave =    -4.54  Std =    21.83
   Min TM value:-264 at an offset of 103.950 usec -> 2021-09-23T17:38:29.909.366.675 (XO: 15.909370267)
   Max TM value:248 at an offset of 106.613 usec -> 2021-09-23T17:38:29.909.369.337 (XO: 15.909372929)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 106.488 usec -> 2021-09-23T17:38:29.909.369.212 (XO: 15.909372804)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 103.775 usec -> 2021-09-23T17:38:29.909.366.500 (XO: 15.909370092)

 - Block number: 3 starting at 2021-09-23T17:38:50.269.732.725 (XO: 36.269734215 - 36.270040216, t=37.2 degC)  Rank: 00076
   TM values: Max =   264  Min =  -263  Ave =    -4.53  Std =    26.67
   Min TM value:-263 at an offset of 92.225 usec -> 2021-09-23T17:38:50.269.824.950 (XO: 36.269826440)
   Max TM value:264 at an offset of 107.363 usec -> 2021-09-23T17:38:50.269.840.087 (XO: 36.269841578)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 107.250 usec -> 2021-09-23T17:38:50.269.839.975 (XO: 36.269841465)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 104.525 usec -> 2021-09-23T17:38:50.269.837.250 (XO: 36.269838740)

 - Block number: 4 starting at 2021-09-23T17:39:08.650.096.725 (XO: 54.650099470 - 54.650405471, t=37.2 degC)  Rank: 00083
   TM values: Max =   248  Min =  -258  Ave =    -4.52  Std =    23.02
   Min TM value:-258 at an offset of 117.813 usec -> 2021-09-23T17:39:08.650.214.537 (XO: 54.650217283)
   Max TM value:248 at an offset of 120.438 usec -> 2021-09-23T17:39:08.650.217.162 (XO: 54.650219908)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 120.325 usec -> 2021-09-23T17:39:08.650.217.050 (XO: 54.650219796)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 123.113 usec -> 2021-09-23T17:39:08.650.219.837 (XO: 54.650222583)
