-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 18 10:25:38 2022 UTC
-- File 20210921_172236_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.231335 to 49.460554
-each separately from 5s to 49s and from 231335us to 934551us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.702638 in 1 packets: 18
      12.342186 in 1 packets: 36
      17.183239 in 1 packets: 54
MIU packet time from 2021-09-21T17:22:44.231.335.000 to 2021-09-21T17:23:28.460.554.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.702638 in 1 packets: 18
      12.342186 in 1 packets: 36
      17.183239 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8448307 to 50.8446846
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999535 in 1 packets: 18
      12.9999569 in 1 packets: 36
      16.9999435 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_172236_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0761669 to 7.0764559
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7025910  -- OK for SL --
   MIU time from 2021-09-21T17:22:44.231.335.000 to 2021-09-21T17:22:44.231.624.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.702638

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7790469 to 21.7793359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3421430  -- OK for SL --
   MIU time from 2021-09-21T17:22:58.934.262.000 to 2021-09-21T17:22:58.934.551.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.342186

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1214789 to 34.1217679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1831830  -- OK for SL --
   MIU time from 2021-09-21T17:23:11.276.737.000 to 2021-09-21T17:23:11.277.026.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.183239

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.3049509 to 51.3052399
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T17:23:28.460.265.000 to 2021-09-21T17:23:28.460.554.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.076166900 corrected by 40bit 10MHz counter of 6.844830700 obtained at 1Hz counter of 5. Result: 5.231336963
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-21T17:22:44.231.194.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T17:23:28.460.413.725
   Min 1st sample XO time among all SL packets: 5.231196688
   Max 1st sample XO time among all SL packets: 49.460416544
   TM values from -269 to 230 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T17:22:44.231.194.725 (XO: 5.231196688 - 5.231502689, t=37.2 degC)  Rank: 00083
   TM values: Max =   213  Min =  -220  Ave =    -4.58  Std =    20.63
   Min TM value:-220 at an offset of 112.538 usec -> 2021-09-21T17:22:44.231.307.262 (XO: 5.231309226)
   Max TM value:213 at an offset of 115.263 usec -> 2021-09-21T17:22:44.231.309.987 (XO: 5.231311951)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 115.113 usec -> 2021-09-21T17:22:44.231.309.837 (XO: 5.231311801)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 117.825 usec -> 2021-09-21T17:22:44.231.312.550 (XO: 5.231314513)

 - Block number: 2 starting at 2021-09-21T17:22:58.934.121.725 (XO: 19.934125508 - 19.934431509, t=37.2 degC)  Rank: 00080
   TM values: Max =   230  Min =  -243  Ave =    -4.54  Std =    24.45
   Min TM value:-243 at an offset of 112.700 usec -> 2021-09-21T17:22:58.934.234.425 (XO: 19.934238208)
   Max TM value:230 at an offset of 115.313 usec -> 2021-09-21T17:22:58.934.237.037 (XO: 19.934240820)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 115.150 usec -> 2021-09-21T17:22:58.934.236.875 (XO: 19.934240658)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 112.488 usec -> 2021-09-21T17:22:58.934.234.212 (XO: 19.934237995)

 - Block number: 3 starting at 2021-09-21T17:23:11.276.596.725 (XO: 32.276598438 - 32.276904439, t=37.2 degC)  Rank: 00075
   TM values: Max =   213  Min =  -253  Ave =    -4.52  Std =    26.15
   Min TM value:-253 at an offset of 112.000 usec -> 2021-09-21T17:23:11.276.708.725 (XO: 32.276710438)
   Max TM value:213 at an offset of 114.550 usec -> 2021-09-21T17:23:11.276.711.275 (XO: 32.276712988)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 114.388 usec -> 2021-09-21T17:23:11.276.711.112 (XO: 32.276712826)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 111.800 usec -> 2021-09-21T17:23:11.276.708.525 (XO: 32.276710238)

 - Block number: 4 starting at 2021-09-21T17:23:28.460.124.725 (XO: 49.460127543 - 49.460433544, t=37.2 degC)  Rank: 00083
   TM values: Max =   218  Min =  -269  Ave =    -4.60  Std =    23.39
   Min TM value:-269 at an offset of 115.038 usec -> 2021-09-21T17:23:28.460.239.762 (XO: 49.460242581)
   Max TM value:218 at an offset of 117.663 usec -> 2021-09-21T17:23:28.460.242.387 (XO: 49.460245206)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 117.450 usec -> 2021-09-21T17:23:28.460.242.175 (XO: 49.460244994)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 114.900 usec -> 2021-09-21T17:23:28.460.239.625 (XO: 49.460242444)
