-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 18 08:18:55 2022 UTC
-- File 20210921_152532_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.245573 to 22.986601
-each separately from 7s to 22s and from 245573us to 986601us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.740450 in 1 packets: 18
MIU packet time from 2021-09-21T15:25:42.245.573.000 to 2021-09-21T15:25:57.986.601.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.740450 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4516913 to 23.4516414
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999501 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_152532_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6972654 to 8.6975544
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7403990  -- OK for SL --
   MIU time from 2021-09-21T15:25:42.245.573.000 to 2021-09-21T15:25:42.245.862.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.740450

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4379534 to 24.4382424
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T15:25:57.986.312.000 to 2021-09-21T15:25:57.986.601.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.697265400 corrected by 40bit 10MHz counter of 8.451691300 obtained at 1Hz counter of 7. Result: 7.245574910
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-21T15:25:42.245.432.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T15:25:57.986.460.725
   Min 1st sample XO time among all SL packets: 7.245434635
   Max 1st sample XO time among all SL packets: 22.986463980
   TM values from -287 to 220 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T15:25:42.245.432.725 (XO: 7.245434635 - 7.245740636, t=37.2 degC)  Rank: 00080
   TM values: Max =   220  Min =  -282  Ave =    -4.66  Std =    22.91
   Min TM value:-282 at an offset of 105.888 usec -> 2021-09-21T15:25:42.245.538.612 (XO: 7.245540523)
   Max TM value:220 at an offset of 105.613 usec -> 2021-09-21T15:25:42.245.538.337 (XO: 7.245540248)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 108.275 usec -> 2021-09-21T15:25:42.245.541.000 (XO: 7.245542910)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 105.725 usec -> 2021-09-21T15:25:42.245.538.450 (XO: 7.245540360)

 - Block number: 2 starting at 2021-09-21T15:25:57.986.171.725 (XO: 22.986174979 - 22.986480980, t=37.2 degC)  Rank: 00082
   TM values: Max =   212  Min =  -287  Ave =    -4.53  Std =    21.20
   Min TM value:-287 at an offset of 120.788 usec -> 2021-09-21T15:25:57.986.292.512 (XO: 22.986295767)
   Max TM value:212 at an offset of 118.488 usec -> 2021-09-21T15:25:57.986.290.212 (XO: 22.986293467)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 123.150 usec -> 2021-09-21T15:25:57.986.294.875 (XO: 22.986298130)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 120.588 usec -> 2021-09-21T15:25:57.986.292.312 (XO: 22.986295567)
