-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 18 01:48:22 2022 UTC
-- File 20210921_111641_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.878488 to 22.905570
-each separately from 0s to 22s and from 878488us to 905570us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.026504 in 1 packets: 18
MIU packet time from 2021-09-21T11:16:44.878.488.000 to 2021-09-21T11:17:06.905.570.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.026504 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4797646 to 23.4796918
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999272 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_111641_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3582536 to 2.3585426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.0264310  -- OK for SL --
   MIU time from 2021-09-21T11:16:44.878.488.000 to 2021-09-21T11:16:44.878.777.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.026504

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3849736 to 24.3852626
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T11:17:06.905.281.000 to 2021-09-21T11:17:06.905.570.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.358253600 corrected by 40bit 10MHz counter of 1.479764600 obtained at 1Hz counter of 0. Result: 0.878491987
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-09-21T11:16:44.878.347.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T11:17:06.905.429.725
   Min 1st sample XO time among all SL packets: 0.878351711
   Max 1st sample XO time among all SL packets: 22.905433603
   TM values from -268 to 232 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T11:16:44.878.347.725 (XO: 0.878351711 - 0.878657712, t=37.2 degC)  Rank: 00086
   TM values: Max =   226  Min =  -246  Ave =    -4.63  Std =    20.82
   Min TM value:-246 at an offset of 120.725 usec -> 2021-09-21T11:16:44.878.468.450 (XO: 0.878472437)
   Max TM value:226 at an offset of 118.475 usec -> 2021-09-21T11:16:44.878.466.200 (XO: 0.878470187)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 123.100 usec -> 2021-09-21T11:16:44.878.470.825 (XO: 0.878474812)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 120.525 usec -> 2021-09-21T11:16:44.878.468.250 (XO: 0.878472237)

 - Block number: 2 starting at 2021-09-21T11:17:06.905.140.725 (XO: 22.905144602 - 22.905450604, t=37.2 degC)  Rank: 00085
   TM values: Max =   232  Min =  -268  Ave =    -4.61  Std =    26.26
   Min TM value:-268 at an offset of 122.138 usec -> 2021-09-21T11:17:06.905.262.862 (XO: 22.905266740)
   Max TM value:232 at an offset of 121.850 usec -> 2021-09-21T11:17:06.905.262.575 (XO: 22.905266453)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 124.525 usec -> 2021-09-21T11:17:06.905.265.250 (XO: 22.905269128)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 121.988 usec -> 2021-09-21T11:17:06.905.262.712 (XO: 22.905266590)
