-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 17 22:51:42 2022 UTC
-- File 20210921_100230_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.798949 to 27.606573
-each separately from 1s to 27s and from 606284us to 799238us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.807046 in 1 packets: 18
MIU packet time from 2021-09-21T10:02:34.798.949.000 to 2021-09-21T10:03:00.606.573.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.807046 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8902032 to 28.8901171
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999139 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_100230_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6891524 to 3.6894414
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8069590  -- OK for SL --
   MIU time from 2021-09-21T10:02:34.798.949.000 to 2021-09-21T10:02:34.799.238.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.807046

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.4964004 to 29.4966894
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T10:03:00.606.284.000 to 2021-09-21T10:03:00.606.573.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.689152400 corrected by 40bit 10MHz counter of 2.890203200 obtained at 1Hz counter of 1. Result: 1.798951837
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-21T10:02:34.798.808.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T10:03:00.606.432.725
   Min 1st sample XO time among all SL packets: 1.798811561
   Max 1st sample XO time among all SL packets: 27.606434087
   TM values from -261 to 205 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T10:02:34.798.808.725 (XO: 1.798811561 - 1.799117562, t=37.2 degC)  Rank: 00080
   TM values: Max =   205  Min =  -261  Ave =    -4.65  Std =    23.54
   Min TM value:-261 at an offset of 157.213 usec -> 2021-09-21T10:02:34.798.965.938 (XO: 1.798968774)
   Max TM value:205 at an offset of 127.863 usec -> 2021-09-21T10:02:34.798.936.587 (XO: 1.798939424)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 159.613 usec -> 2021-09-21T10:02:34.798.968.338 (XO: 1.798971174)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 129.900 usec -> 2021-09-21T10:02:34.798.938.625 (XO: 1.798941462)

 - Block number: 2 starting at 2021-09-21T10:03:00.606.143.725 (XO: 27.606145086 - 27.606451087, t=37.2 degC)  Rank: 00083
   TM values: Max =   202  Min =  -261  Ave =    -4.67  Std =    28.30
   Min TM value:-261 at an offset of 159.751 usec -> 2021-09-21T10:03:00.606.303.475 (XO: 27.606304836)
   Max TM value:202 at an offset of 118.138 usec -> 2021-09-21T10:03:00.606.261.862 (XO: 27.606263224)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 117.950 usec -> 2021-09-21T10:03:00.606.261.675 (XO: 27.606263036)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 187.038 usec -> 2021-09-21T10:03:00.606.330.763 (XO: 27.606332124)
