-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 17 12:21:57 2022 UTC
-- File 20210921_072756_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.363388 to 47.662478
-each separately from 13s to 47s and from 262708us to 662478us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.029283 in 1 packets: 18
      5.869459 in 1 packets: 36
      17.399192 in 1 packets: 54
MIU packet time from 2021-09-21T07:28:12.363.388.000 to 2021-09-21T07:28:46.662.478.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.029283 in 1 packets: 18
      5.869459 in 1 packets: 36
      17.399192 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9140921 to 48.9139791
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999634 in 1 packets: 18
      5.9999800 in 1 packets: 36
      16.9999436 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      04 -> 49.8 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_072756_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.2774805 to 15.2777695
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0292470  -- OK for SL --
   MIU time from 2021-09-21T07:28:12.363.388.000 to 2021-09-21T07:28:12.363.677.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.029283

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3070165 to 26.3073055
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.8694390  -- OK for SL --
   MIU time from 2021-09-21T07:28:23.392.960.000 to 2021-09-21T07:28:23.393.249.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.869459

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.1767445 to 32.1770335
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3991350  -- OK for SL --
   MIU time from 2021-09-21T07:28:29.262.708.000 to 2021-09-21T07:28:29.262.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.399192

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.5761685 to 49.5764575
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T07:28:46.662.189.000 to 2021-09-21T07:28:46.662.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.277480500 corrected by 40bit 10MHz counter of 14.914092100 obtained at 1Hz counter of 13. Result: 13.363389636
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-21T07:28:12.363.247.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T07:28:46.662.337.725
   Min 1st sample XO time among all SL packets: 13.363249360
   Max 1st sample XO time among all SL packets: 47.662340377
   TM values from -263 to 256 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T07:28:12.363.247.725 (XO: 13.363249360 - 13.363555361, t=37.2 degC)  Rank: 00082
   TM values: Max =   204  Min =  -241  Ave =    -4.52  Std =    20.73
   Min TM value:-241 at an offset of 102.863 usec -> 2021-09-21T07:28:12.363.350.587 (XO: 13.363352223)
   Max TM value:204 at an offset of 105.500 usec -> 2021-09-21T07:28:12.363.353.225 (XO: 13.363354860)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 105.350 usec -> 2021-09-21T07:28:12.363.353.075 (XO: 13.363354710)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 102.663 usec -> 2021-09-21T07:28:12.363.350.387 (XO: 13.363352023)

 - Block number: 2 starting at 2021-09-21T07:28:23.392.819.725 (XO: 24.392822061 - 24.393128062, t=37.2 degC)  Rank: 00076
   TM values: Max =   256  Min =  -248  Ave =    -4.52  Std =    27.42
   Min TM value:-248 at an offset of 119.863 usec -> 2021-09-21T07:28:23.392.939.587 (XO: 24.392941924)
   Max TM value:256 at an offset of 119.600 usec -> 2021-09-21T07:28:23.392.939.325 (XO: 24.392941661)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 117.588 usec -> 2021-09-21T07:28:23.392.937.312 (XO: 24.392939648)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 119.713 usec -> 2021-09-21T07:28:23.392.939.437 (XO: 24.392941774)

 - Block number: 3 starting at 2021-09-21T07:28:29.262.567.725 (XO: 30.262569591 - 30.262875592, t=37.2 degC)  Rank: 00082
   TM values: Max =   233  Min =  -244  Ave =    -4.46  Std =    27.83
   Min TM value:-244 at an offset of 185.563 usec -> 2021-09-21T07:28:29.262.753.288 (XO: 30.262755155)
   Max TM value:233 at an offset of 128.938 usec -> 2021-09-21T07:28:29.262.696.662 (XO: 30.262698529)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 131.600 usec -> 2021-09-21T07:28:29.262.699.325 (XO: 30.262701192)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 129.050 usec -> 2021-09-21T07:28:29.262.696.775 (XO: 30.262698642)

 - Block number: 4 starting at 2021-09-21T07:28:46.662.048.725 (XO: 47.662051376 - 47.662357377, t=37.2 degC)  Rank: 00080
   TM values: Max =   220  Min =  -263  Ave =    -4.52  Std =    22.29
   Min TM value:-263 at an offset of 116.438 usec -> 2021-09-21T07:28:46.662.165.162 (XO: 47.662167814)
   Max TM value:220 at an offset of 116.150 usec -> 2021-09-21T07:28:46.662.164.875 (XO: 47.662167526)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 118.800 usec -> 2021-09-21T07:28:46.662.167.525 (XO: 47.662170176)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 116.275 usec -> 2021-09-21T07:28:46.662.165.000 (XO: 47.662167651)
