-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 17 07:22:17 2022 UTC
-- File 20210921_031511_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.156920 to 67.868566
-each separately from 10s to 67s and from 37129us to 868566us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.879920 in 1 packets: 18
      15.040080 in 1 packets: 36
      19.700223 in 1 packets: 54
      13.089978 in 1 packets: 72
MIU packet time from 2021-09-21T03:15:24.156.920.000 to 2021-09-21T03:16:21.868.566.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.879920 in 1 packets: 18
      15.040080 in 1 packets: 36
      19.700223 in 1 packets: 54
      13.089978 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9596858 to 68.9594969
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
      14.9999504 in 1 packets: 36
      18.9999370 in 1 packets: 54
      12.9999568 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210921_031511_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1166066 to 12.1168956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.8798870  -- OK for SL --
   MIU time from 2021-09-21T03:15:24.156.920.000 to 2021-09-21T03:15:24.157.209.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.879920

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9967826 to 21.9970716
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0400310  -- OK for SL --
   MIU time from 2021-09-21T03:15:34.037.129.000 to 2021-09-21T03:15:34.037.418.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.040080

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.0371026 to 37.0373916
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7001590  -- OK for SL --
   MIU time from 2021-09-21T03:15:49.077.498.000 to 2021-09-21T03:15:49.077.787.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.700223

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.7375506 to 56.7378396
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0899350  -- OK for SL --
   MIU time from 2021-09-21T03:16:08.778.010.000 to 2021-09-21T03:16:08.778.299.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.089978

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.8277746 to 69.8280636
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-21T03:16:21.868.277.000 to 2021-09-21T03:16:21.868.566.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.116606600 corrected by 40bit 10MHz counter of 11.959685800 obtained at 1Hz counter of 10. Result: 10.156921318
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-21T03:15:24.156.779.725
   Max 1st sample MIU time among all SL packets: 2021-09-21T03:16:21.868.425.725
   Min 1st sample XO time among all SL packets: 10.156781042
   Max 1st sample XO time among all SL packets: 67.868429291
   TM values from -261 to 239 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-21T03:15:24.156.779.725 (XO: 10.156781042 - 10.157087043, t=37.2 degC)  Rank: 00089
   TM values: Max =   239  Min =  -254  Ave =    -4.57  Std =    18.80
   Min TM value:-254 at an offset of 128.513 usec -> 2021-09-21T03:15:24.156.908.237 (XO: 10.156909555)
   Max TM value:239 at an offset of 126.388 usec -> 2021-09-21T03:15:24.156.906.112 (XO: 10.156907430)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 126.275 usec -> 2021-09-21T03:15:24.156.906.000 (XO: 10.156907318)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 128.375 usec -> 2021-09-21T03:15:24.156.908.100 (XO: 10.156909418)

 - Block number: 2 starting at 2021-09-21T03:15:34.036.988.725 (XO: 20.036989747 - 20.037295748, t=37.2 degC)  Rank: 00079
   TM values: Max =   207  Min =  -219  Ave =    -4.52  Std =    17.37
   Min TM value:-219 at an offset of 130.113 usec -> 2021-09-21T03:15:34.037.118.837 (XO: 20.037119860)
   Max TM value:207 at an offset of 127.913 usec -> 2021-09-21T03:15:34.037.116.637 (XO: 20.037117660)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 127.788 usec -> 2021-09-21T03:15:34.037.116.512 (XO: 20.037117535)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 129.950 usec -> 2021-09-21T03:15:34.037.118.675 (XO: 20.037119697)

 - Block number: 3 starting at 2021-09-21T03:15:49.077.357.725 (XO: 35.077359480 - 35.077665481, t=37.2 degC)  Rank: 00080
   TM values: Max =   195  Min =  -247  Ave =    -4.50  Std =    20.86
   Min TM value:-247 at an offset of 130.900 usec -> 2021-09-21T03:15:49.077.488.625 (XO: 35.077490381)
   Max TM value:195 at an offset of 133.463 usec -> 2021-09-21T03:15:49.077.491.187 (XO: 35.077492943)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 133.313 usec -> 2021-09-21T03:15:49.077.491.037 (XO: 35.077492793)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 130.713 usec -> 2021-09-21T03:15:49.077.488.437 (XO: 35.077490193)

 - Block number: 4 starting at 2021-09-21T03:16:08.777.869.725 (XO: 54.777872792 - 54.778178793, t=37.2 degC)  Rank: 00078
   TM values: Max =   205  Min =  -235  Ave =    -4.57  Std =    19.72
   Min TM value:-235 at an offset of 103.850 usec -> 2021-09-21T03:16:08.777.973.575 (XO: 54.777976642)
   Max TM value:205 at an offset of 106.463 usec -> 2021-09-21T03:16:08.777.976.187 (XO: 54.777979255)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 106.350 usec -> 2021-09-21T03:16:08.777.976.075 (XO: 54.777979142)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 103.675 usec -> 2021-09-21T03:16:08.777.973.400 (XO: 54.777976467)

 - Block number: 5 starting at 2021-09-21T03:16:21.868.136.725 (XO: 67.868140290 - 67.868446291, t=37.2 degC)  Rank: 00082
   TM values: Max =   204  Min =  -261  Ave =    -4.56  Std =    21.19
   Min TM value:-261 at an offset of 117.075 usec -> 2021-09-21T03:16:21.868.253.800 (XO: 67.868257365)
   Max TM value:204 at an offset of 116.788 usec -> 2021-09-21T03:16:21.868.253.512 (XO: 67.868257078)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 119.450 usec -> 2021-09-21T03:16:21.868.256.175 (XO: 67.868259740)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 116.888 usec -> 2021-09-21T03:16:21.868.253.612 (XO: 67.868257178)
