-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 18:01:22 2022 UTC
-- File 20210920_144019_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.732127 to 20.332148
-each separately from 9s to 20s and from 331859us to 732416us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.599443 in 1 packets: 18
MIU packet time from 2021-09-20T14:40:31.732.127.000 to 2021-09-20T14:40:42.332.148.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.599443 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9053223 to 21.9052858
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999635 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_144019_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6374501 to 11.6377391
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5994070  -- OK for SL --
   MIU time from 2021-09-20T14:40:31.732.127.000 to 2021-09-20T14:40:31.732.416.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.599443

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.2371461 to 22.2374351
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T14:40:42.331.859.000 to 2021-09-20T14:40:42.332.148.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.637450100 corrected by 40bit 10MHz counter of 10.905322300 obtained at 1Hz counter of 9. Result: 9.732130216
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T14:40:31.731.986.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T14:40:42.332.007.725
   Min 1st sample XO time among all SL packets: 9.731989941
   Max 1st sample XO time among all SL packets: 20.332010121
   TM values from -241 to 207 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T14:40:31.731.986.725 (XO: 9.731989941 - 9.732295942, t=37.2 degC)  Rank: 00081
   TM values: Max =   207  Min =  -241  Ave =    -4.52  Std =    20.12
   Min TM value:-241 at an offset of 125.450 usec -> 2021-09-20T14:40:31.732.112.175 (XO: 9.732115391)
   Max TM value:207 at an offset of 125.163 usec -> 2021-09-20T14:40:31.732.111.887 (XO: 9.732115103)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 127.813 usec -> 2021-09-20T14:40:31.732.114.537 (XO: 9.732117753)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 125.275 usec -> 2021-09-20T14:40:31.732.112.000 (XO: 9.732115216)

 - Block number: 2 starting at 2021-09-20T14:40:42.331.718.725 (XO: 20.331721120 - 20.332027121, t=37.2 degC)  Rank: 00080
   TM values: Max =   180  Min =  -237  Ave =    -4.50  Std =    19.94
   Min TM value:-237 at an offset of 128.138 usec -> 2021-09-20T14:40:42.331.846.862 (XO: 20.331849258)
   Max TM value:180 at an offset of 127.838 usec -> 2021-09-20T14:40:42.331.846.562 (XO: 20.331848958)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 130.613 usec -> 2021-09-20T14:40:42.331.849.337 (XO: 20.331851733)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 127.950 usec -> 2021-09-20T14:40:42.331.846.675 (XO: 20.331849070)
