-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 15:48:34 2022 UTC
-- File 20210920_131257_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.865718 to 16.306126
-each separately from 4s to 16s and from 305837us to 866007us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.439830 in 1 packets: 18
MIU packet time from 2021-09-20T13:13:04.865.718.000 to 2021-09-20T13:13:16.306.126.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.439830 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8590771 to 17.8590376
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999605 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_131257_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7247959 to 6.7250849
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.4397910  -- OK for SL --
   MIU time from 2021-09-20T13:13:04.865.718.000 to 2021-09-20T13:13:04.866.007.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.439830

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.1648759 to 18.1651649
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T13:13:16.305.837.000 to 2021-09-20T13:13:16.306.126.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.724795900 corrected by 40bit 10MHz counter of 5.859077100 obtained at 1Hz counter of 4. Result: 4.865721657
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T13:13:04.865.577.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T13:13:16.305.985.725
   Min 1st sample XO time among all SL packets: 4.865581381
   Max 1st sample XO time among all SL packets: 16.305988035
   TM values from -252 to 200 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T13:13:04.865.577.725 (XO: 4.865581381 - 4.865887382, t=37.2 degC)  Rank: 00087
   TM values: Max =   200  Min =  -228  Ave =    -4.52  Std =    19.30
   Min TM value:-228 at an offset of 123.713 usec -> 2021-09-20T13:13:04.865.701.437 (XO: 4.865705094)
   Max TM value:200 at an offset of 126.400 usec -> 2021-09-20T13:13:04.865.704.125 (XO: 4.865707782)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 126.250 usec -> 2021-09-20T13:13:04.865.703.975 (XO: 4.865707632)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 123.538 usec -> 2021-09-20T13:13:04.865.701.262 (XO: 4.865704919)

 - Block number: 2 starting at 2021-09-20T13:13:16.305.696.725 (XO: 16.305699034 - 16.306005035, t=37.2 degC)  Rank: 00078
   TM values: Max =   196  Min =  -252  Ave =    -4.61  Std =    30.15
   Min TM value:-252 at an offset of 183.588 usec -> 2021-09-20T13:13:16.305.880.313 (XO: 16.305882622)
   Max TM value:196 at an offset of 111.750 usec -> 2021-09-20T13:13:16.305.808.475 (XO: 16.305810784)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 116.400 usec -> 2021-09-20T13:13:16.305.813.125 (XO: 16.305815434)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 113.825 usec -> 2021-09-20T13:13:16.305.810.550 (XO: 16.305812859)
