-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 14:59:37 2022 UTC
-- File 20210920_123746_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.215299 to 27.085906
-each separately from 13s to 27s and from 85617us to 215588us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.870029 in 1 packets: 18
MIU packet time from 2021-09-20T12:38:02.215.299.000 to 2021-09-20T12:38:16.085.906.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.870029 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8907612 to 28.8907149
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999537 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_123746_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1060608 to 15.1063498
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8699830  -- OK for SL --
   MIU time from 2021-09-20T12:38:02.215.299.000 to 2021-09-20T12:38:02.215.588.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.870029

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.9763328 to 28.9766218
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T12:38:16.085.617.000 to 2021-09-20T12:38:16.085.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.106060800 corrected by 40bit 10MHz counter of 14.890761200 obtained at 1Hz counter of 13. Result: 13.215300310
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-20T12:38:02.215.158.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T12:38:16.085.765.725
   Min 1st sample XO time among all SL packets: 13.215160035
   Max 1st sample XO time among all SL packets: 27.085766899
   TM values from -292 to 249 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T12:38:02.215.158.725 (XO: 13.215160035 - 13.215466036, t=37.2 degC)  Rank: 00077
   TM values: Max =   236  Min =  -252  Ave =    -4.57  Std =    28.36
   Min TM value:-252 at an offset of 177.751 usec -> 2021-09-20T12:38:02.215.336.475 (XO: 13.215337786)
   Max TM value:236 at an offset of 119.513 usec -> 2021-09-20T12:38:02.215.278.237 (XO: 13.215279548)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 117.438 usec -> 2021-09-20T12:38:02.215.276.162 (XO: 13.215277473)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 177.613 usec -> 2021-09-20T12:38:02.215.336.338 (XO: 13.215337648)

 - Block number: 2 starting at 2021-09-20T12:38:16.085.476.725 (XO: 27.085477899 - 27.085783900, t=37.2 degC)  Rank: 00085
   TM values: Max =   249  Min =  -292  Ave =    -4.62  Std =    33.07
   Min TM value:-292 at an offset of 106.513 usec -> 2021-09-20T12:38:16.085.583.237 (XO: 27.085584411)
   Max TM value:249 at an offset of 148.975 usec -> 2021-09-20T12:38:16.085.625.700 (XO: 27.085626874)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 151.588 usec -> 2021-09-20T12:38:16.085.628.312 (XO: 27.085629487)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 106.375 usec -> 2021-09-20T12:38:16.085.583.100 (XO: 27.085584274)
