-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 13:47:46 2022 UTC
-- File 20210920_113548_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.374201 to 27.034219
-each separately from 12s to 27s and from 33930us to 374490us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.659440 in 1 packets: 18
MIU packet time from 2021-09-20T11:36:03.374.201.000 to 2021-09-20T11:36:18.034.219.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.659440 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9238936 to 28.9238442
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999506 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_113548_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.2980952 to 14.2983842
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6593910  -- OK for SL --
   MIU time from 2021-09-20T11:36:03.374.201.000 to 2021-09-20T11:36:03.374.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.659440

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.9577752 to 28.9580642
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T11:36:18.033.930.000 to 2021-09-20T11:36:18.034.219.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.298095200 corrected by 40bit 10MHz counter of 13.923893600 obtained at 1Hz counter of 12. Result: 12.374202835
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T11:36:03.374.060.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T11:36:18.034.078.725
   Min 1st sample XO time among all SL packets: 12.374062559
   Max 1st sample XO time among all SL packets: 27.034079837
   TM values from -235 to 210 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T11:36:03.374.060.725 (XO: 12.374062559 - 12.374368560, t=37.2 degC)  Rank: 00091
   TM values: Max =   210  Min =  -222  Ave =    -4.62  Std =    21.98
   Min TM value:-222 at an offset of 120.050 usec -> 2021-09-20T11:36:03.374.180.775 (XO: 12.374182610)
   Max TM value:210 at an offset of 122.713 usec -> 2021-09-20T11:36:03.374.183.437 (XO: 12.374185272)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 122.550 usec -> 2021-09-20T11:36:03.374.183.275 (XO: 12.374185110)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 119.850 usec -> 2021-09-20T11:36:03.374.180.575 (XO: 12.374182410)

 - Block number: 2 starting at 2021-09-20T11:36:18.033.789.725 (XO: 27.033790837 - 27.034096838, t=37.2 degC)  Rank: 00079
   TM values: Max =   186  Min =  -235  Ave =    -4.58  Std =    20.63
   Min TM value:-235 at an offset of 92.788 usec -> 2021-09-20T11:36:18.033.882.512 (XO: 27.033883624)
   Max TM value:186 at an offset of 121.650 usec -> 2021-09-20T11:36:18.033.911.375 (XO: 27.033912487)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 121.475 usec -> 2021-09-20T11:36:18.033.911.200 (XO: 27.033912312)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 92.613 usec -> 2021-09-20T11:36:18.033.882.337 (XO: 27.033883449)
