-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 04:07:28 2022 UTC
-- File 20210920_034141_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.874923 to 38.332966
-each separately from 1s to 38s and from 332677us to 875212us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.718779 in 1 packets: 18
      18.738397 in 1 packets: 36
MIU packet time from 2021-09-20T03:41:45.874.923.000 to 2021-09-20T03:42:22.332.966.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.718779 in 1 packets: 18
      18.738397 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4431550 to 39.4430325
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999405 in 1 packets: 18
      18.9999370 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_034141_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3180788 to 3.3183678
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7187190  -- OK for SL --
   MIU time from 2021-09-20T03:41:45.874.923.000 to 2021-09-20T03:41:45.875.212.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.718779

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0370868 to 21.0373758
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7383350  -- OK for SL --
   MIU time from 2021-09-20T03:42:03.593.991.000 to 2021-09-20T03:42:03.594.280.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.738397

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.7757108 to 39.7759998
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T03:42:22.332.677.000 to 2021-09-20T03:42:22.332.966.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.318078800 corrected by 40bit 10MHz counter of 2.443155000 obtained at 1Hz counter of 1. Result: 1.874926687
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T03:41:45.874.782.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T03:42:22.332.825.725
   Min 1st sample XO time among all SL packets: 1.874786412
   Max 1st sample XO time among all SL packets: 38.332828123
   TM values from -264 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T03:41:45.874.782.725 (XO: 1.874786412 - 1.875092413, t=37.2 degC)  Rank: 00086
   TM values: Max =   209  Min =  -242  Ave =    -4.66  Std =    19.30
   Min TM value:-242 at an offset of 124.425 usec -> 2021-09-20T03:41:45.874.907.150 (XO: 1.874910837)
   Max TM value:209 at an offset of 126.975 usec -> 2021-09-20T03:41:45.874.909.700 (XO: 1.874913387)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 126.800 usec -> 2021-09-20T03:41:45.874.909.525 (XO: 1.874913212)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 124.238 usec -> 2021-09-20T03:41:45.874.906.962 (XO: 1.874910650)

 - Block number: 2 starting at 2021-09-20T03:42:03.593.850.725 (XO: 19.593852985 - 19.594158986, t=37.2 degC)  Rank: 00077
   TM values: Max =   209  Min =  -264  Ave =    -4.70  Std =    19.94
   Min TM value:-264 at an offset of 115.088 usec -> 2021-09-20T03:42:03.593.965.812 (XO: 19.593968073)
   Max TM value:209 at an offset of 114.775 usec -> 2021-09-20T03:42:03.593.965.500 (XO: 19.593967760)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 117.450 usec -> 2021-09-20T03:42:03.593.968.175 (XO: 19.593970435)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 114.875 usec -> 2021-09-20T03:42:03.593.965.600 (XO: 19.593967860)

 - Block number: 3 starting at 2021-09-20T03:42:22.332.536.725 (XO: 38.332539122 - 38.332845123, t=37.2 degC)  Rank: 00077
   TM values: Max =   179  Min =  -203  Ave =    -4.66  Std =    17.40
   Min TM value:-203 at an offset of 124.425 usec -> 2021-09-20T03:42:22.332.661.150 (XO: 38.332663548)
   Max TM value:179 at an offset of 127.238 usec -> 2021-09-20T03:42:22.332.663.962 (XO: 38.332666360)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 127.075 usec -> 2021-09-20T03:42:22.332.663.800 (XO: 38.332666198)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 124.213 usec -> 2021-09-20T03:42:22.332.660.937 (XO: 38.332663335)
