-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 02:01:03 2022 UTC
-- File 20210920_022901_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.735892 to 31.576228
-each separately from 7s to 31s and from 255982us to 736181us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.519801 in 1 packets: 18
      16.319668 in 1 packets: 36
MIU packet time from 2021-09-20T02:29:11.735.892.000 to 2021-09-20T02:29:35.576.228.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.519801 in 1 packets: 18
      16.319668 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9217770 to 32.9216976
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999735 in 1 packets: 18
      15.9999471 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_022901_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6576693 to 9.6579583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.5197750  -- OK for SL --
   MIU time from 2021-09-20T02:29:11.735.892.000 to 2021-09-20T02:29:11.736.181.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.519801

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1777333 to 17.1780223
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3196150  -- OK for SL --
   MIU time from 2021-09-20T02:29:19.255.982.000 to 2021-09-20T02:29:19.256.271.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.319668

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.4976373 to 33.4979263
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T02:29:35.575.939.000 to 2021-09-20T02:29:35.576.228.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.657669300 corrected by 40bit 10MHz counter of 8.921777000 obtained at 1Hz counter of 7. Result: 7.735894802
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T02:29:11.735.751.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T02:29:35.576.087.725
   Min 1st sample XO time among all SL packets: 7.735754527
   Max 1st sample XO time among all SL packets: 31.576090326
   TM values from -249 to 202 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T02:29:11.735.751.725 (XO: 7.735754527 - 7.736060528, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -248  Ave =    -4.47  Std =    20.37
   Min TM value:-248 at an offset of 121.800 usec -> 2021-09-20T02:29:11.735.873.525 (XO: 7.735876327)
   Max TM value:192 at an offset of 124.438 usec -> 2021-09-20T02:29:11.735.876.162 (XO: 7.735878964)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 124.263 usec -> 2021-09-20T02:29:11.735.875.987 (XO: 7.735878789)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 121.638 usec -> 2021-09-20T02:29:11.735.873.362 (XO: 7.735876164)

 - Block number: 2 starting at 2021-09-20T02:29:19.255.841.725 (XO: 15.255843369 - 15.256149370, t=37.2 degC)  Rank: 00080
   TM values: Max =   202  Min =  -249  Ave =    -4.58  Std =    19.82
   Min TM value:-249 at an offset of 109.725 usec -> 2021-09-20T02:29:19.255.951.450 (XO: 15.255953095)
   Max TM value:202 at an offset of 109.413 usec -> 2021-09-20T02:29:19.255.951.137 (XO: 15.255952782)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 112.163 usec -> 2021-09-20T02:29:19.255.953.887 (XO: 15.255955532)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 109.550 usec -> 2021-09-20T02:29:19.255.951.275 (XO: 15.255952920)

 - Block number: 3 starting at 2021-09-20T02:29:35.575.798.725 (XO: 31.575801325 - 31.576107326, t=37.2 degC)  Rank: 00082
   TM values: Max =   200  Min =  -236  Ave =    -4.46  Std =    30.39
   Min TM value:-236 at an offset of 110.063 usec -> 2021-09-20T02:29:35.575.908.787 (XO: 31.575911388)
   Max TM value:200 at an offset of 158.601 usec -> 2021-09-20T02:29:35.575.957.325 (XO: 31.575959926)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 158.451 usec -> 2021-09-20T02:29:35.575.957.175 (XO: 31.575959776)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 141.225 usec -> 2021-09-20T02:29:35.575.939.950 (XO: 31.575942551)
