-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 16 01:41:57 2022 UTC
-- File 20210920_021801_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.404164 to 58.382685
-each separately from 7s to 58s and from 382396us to 833045us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.099219 in 1 packets: 18
      23.328795 in 1 packets: 36
      16.549351 in 1 packets: 54
MIU packet time from 2021-09-20T02:18:11.404.164.000 to 2021-09-20T02:19:02.382.685.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.099219 in 1 packets: 18
      23.328795 in 1 packets: 36
      16.549351 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8617641 to 59.8615958
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999637 in 1 packets: 18
      22.9999240 in 1 packets: 36
      16.9999440 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      3 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      4 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      3 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      4 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_021801_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2659286 to 9.2662176
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0991830  -- OK for SL --
   MIU time from 2021-09-20T02:18:11.404.164.000 to 2021-09-20T02:18:11.404.453.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.099219

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3654006 to 20.3656896
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3287190  -- OK for SL --
   MIU time from 2021-09-20T02:18:22.503.672.000 to 2021-09-20T02:18:22.503.961.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.328795

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6944086 to 43.6946976
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5492950  -- OK for SL --
   MIU time from 2021-09-20T02:18:45.832.756.000 to 2021-09-20T02:18:45.833.045.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.549351

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.2439926 to 60.2442816
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T02:19:02.382.396.000 to 2021-09-20T02:19:02.382.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.265928600 corrected by 40bit 10MHz counter of 8.861764100 obtained at 1Hz counter of 7. Result: 7.404165834
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-20T02:18:11.404.023.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T02:19:02.382.544.725
   Min 1st sample XO time among all SL packets: 7.404025558
   Max 1st sample XO time among all SL packets: 58.382546787
   TM values from -256 to 212 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T02:18:11.404.023.725 (XO: 7.404025558 - 7.404331559, t=37.2 degC)  Rank: 00078
   TM values: Max =   212  Min =  -249  Ave =    -4.58  Std =    20.99
   Min TM value:-249 at an offset of 111.813 usec -> 2021-09-20T02:18:11.404.135.537 (XO: 7.404137371)
   Max TM value:212 at an offset of 111.500 usec -> 2021-09-20T02:18:11.404.135.225 (XO: 7.404137059)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 114.200 usec -> 2021-09-20T02:18:11.404.137.925 (XO: 7.404139759)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 111.600 usec -> 2021-09-20T02:18:11.404.135.325 (XO: 7.404137159)

 - Block number: 2 starting at 2021-09-20T02:18:22.503.531.725 (XO: 18.503534237 - 18.503840238, t=37.2 degC)  Rank: 00086
   TM values: Max =   210  Min =  -245  Ave =    -4.63  Std =    20.65
   Min TM value:-245 at an offset of 111.100 usec -> 2021-09-20T02:18:22.503.642.825 (XO: 18.503645337)
   Max TM value:210 at an offset of 110.813 usec -> 2021-09-20T02:18:22.503.642.537 (XO: 18.503645050)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 113.525 usec -> 2021-09-20T02:18:22.503.645.250 (XO: 18.503647762)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 110.925 usec -> 2021-09-20T02:18:22.503.642.650 (XO: 18.503645162)

 - Block number: 3 starting at 2021-09-20T02:18:45.832.615.725 (XO: 41.832619273 - 41.832925274, t=37.2 degC)  Rank: 00082
   TM values: Max =   195  Min =  -256  Ave =    -4.54  Std =    20.13
   Min TM value:-256 at an offset of 125.025 usec -> 2021-09-20T02:18:45.832.740.750 (XO: 41.832744298)
   Max TM value:195 at an offset of 124.725 usec -> 2021-09-20T02:18:45.832.740.450 (XO: 41.832743998)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 127.450 usec -> 2021-09-20T02:18:45.832.743.175 (XO: 41.832746723)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 124.888 usec -> 2021-09-20T02:18:45.832.740.612 (XO: 41.832744161)

 - Block number: 4 starting at 2021-09-20T02:19:02.382.255.725 (XO: 58.382257786 - 58.382563787, t=37.2 degC)  Rank: 00084
   TM values: Max =   199  Min =  -217  Ave =    -4.46  Std =    18.79
   Min TM value:-217 at an offset of 122.163 usec -> 2021-09-20T02:19:02.382.377.887 (XO: 58.382379949)
   Max TM value:199 at an offset of 124.850 usec -> 2021-09-20T02:19:02.382.380.575 (XO: 58.382382637)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 124.688 usec -> 2021-09-20T02:19:02.382.380.412 (XO: 58.382382474)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 121.950 usec -> 2021-09-20T02:19:02.382.377.675 (XO: 58.382379737)
