-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 15 23:16:11 2022 UTC
-- File 20210920_003032_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.605639 to 21.594302
-each separately from 5s to 21s and from 594013us to 605928us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.988085 in 1 packets: 18
MIU packet time from 2021-09-20T00:30:40.605.639.000 to 2021-09-20T00:30:56.594.302.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.988085 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9064311 to 22.9063780
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999469 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210920_003032_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.5120713 to 7.5123603
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9880310  -- OK for SL --
   MIU time from 2021-09-20T00:30:40.605.639.000 to 2021-09-20T00:30:40.605.928.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.988085

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.5003913 to 23.5006803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-20T00:30:56.594.013.000 to 2021-09-20T00:30:56.594.302.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.512071300 corrected by 40bit 10MHz counter of 6.906431100 obtained at 1Hz counter of 5. Result: 5.605642259
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-09-20T00:30:40.605.498.725
   Max 1st sample MIU time among all SL packets: 2021-09-20T00:30:56.594.161.725
   Min 1st sample XO time among all SL packets: 5.605501984
   Max 1st sample XO time among all SL packets: 21.594164045
   TM values from -256 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-20T00:30:40.605.498.725 (XO: 5.605501984 - 5.605807985, t=37.2 degC)  Rank: 00083
   TM values: Max =   217  Min =  -221  Ave =    -4.58  Std =    20.24
   Min TM value:-221 at an offset of 120.050 usec -> 2021-09-20T00:30:40.605.618.775 (XO: 5.605622034)
   Max TM value:217 at an offset of 122.750 usec -> 2021-09-20T00:30:40.605.621.475 (XO: 5.605624734)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 122.588 usec -> 2021-09-20T00:30:40.605.621.312 (XO: 5.605624572)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 119.813 usec -> 2021-09-20T00:30:40.605.618.537 (XO: 5.605621797)

 - Block number: 2 starting at 2021-09-20T00:30:56.593.872.725 (XO: 21.593875044 - 21.594181045, t=37.2 degC)  Rank: 00080
   TM values: Max =   238  Min =  -256  Ave =    -4.51  Std =    26.06
   Min TM value:-256 at an offset of 126.713 usec -> 2021-09-20T00:30:56.593.999.437 (XO: 21.594001757)
   Max TM value:238 at an offset of 96.038 usec -> 2021-09-20T00:30:56.593.968.762 (XO: 21.593971082)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 124.313 usec -> 2021-09-20T00:30:56.593.997.037 (XO: 21.593999357)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 98.038 usec -> 2021-09-20T00:30:56.593.970.762 (XO: 21.593973082)
