-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 13 05:36:30 2022 UTC
-- File 20210917_013936_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.065361 to 64.094947
-each separately from 1s to 64s and from 65361us to 495576us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.429637 in 1 packets: 18
      23.659470 in 1 packets: 36
      7.988426 in 1 packets: 54
      14.950608 in 1 packets: 72
MIU packet time from 2021-09-17T01:39:40.065.361.000 to 2021-09-17T01:40:43.094.947.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.429637 in 1 packets: 18
      23.659470 in 1 packets: 36
      7.988426 in 1 packets: 54
      14.950608 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7565520 to 65.7563429
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999468 in 1 packets: 18
      23.9999203 in 1 packets: 36
      7.9999735 in 1 packets: 54
      14.9999503 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      03 -> 51.1 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210917_013936_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8219143 to 2.8222033
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4295830  -- OK for SL --
   MIU time from 2021-09-17T01:39:40.065.361.000 to 2021-09-17T01:39:40.065.650.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.429637

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2517863 to 19.2520753
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.6593910  -- OK for SL --
   MIU time from 2021-09-17T01:39:56.495.287.000 to 2021-09-17T01:39:56.495.576.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.659470

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.9114663 to 42.9117553
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9883990  -- OK for SL --
   MIU time from 2021-09-17T01:40:20.155.046.000 to 2021-09-17T01:40:20.155.335.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.988426

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.9001543 to 50.9004433
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9505590  -- OK for SL --
   MIU time from 2021-09-17T01:40:28.143.761.000 to 2021-09-17T01:40:28.144.050.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.950608

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.8510023 to 65.8512913
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-17T01:40:43.094.658.000 to 2021-09-17T01:40:43.094.947.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.821914300 corrected by 40bit 10MHz counter of 2.756552000 obtained at 1Hz counter of 1. Result: 1.065362522
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-17T01:39:40.065.220.725
   Max 1st sample MIU time among all SL packets: 2021-09-17T01:40:43.094.806.725
   Min 1st sample XO time among all SL packets: 1.065222247
   Max 1st sample XO time among all SL packets: 64.094808447
   TM values from -223 to 187 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-17T01:39:40.065.220.725 (XO: 1.065222247 - 1.065528248, t=37.2 degC)  Rank: 00072
   TM values: Max =   187  Min =  -223  Ave =    -4.48  Std =    21.45
   Min TM value:-223 at an offset of 117.463 usec -> 2021-09-17T01:39:40.065.338.187 (XO: 1.065339710)
   Max TM value:187 at an offset of 117.138 usec -> 2021-09-17T01:39:40.065.337.862 (XO: 1.065339385)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 115.163 usec -> 2021-09-17T01:39:40.065.335.887 (XO: 1.065337410)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 117.300 usec -> 2021-09-17T01:39:40.065.338.025 (XO: 1.065339547)

 - Block number: 2 starting at 2021-09-17T01:39:56.495.146.725 (XO: 17.495148909 - 17.495454910, t=37.2 degC)  Rank: 00065
   TM values: Max =   176  Min =  -210  Ave =    -4.56  Std =    17.38
   Min TM value:-210 at an offset of 127.438 usec -> 2021-09-17T01:39:56.495.274.162 (XO: 17.495276346)
   Max TM value:176 at an offset of 130.038 usec -> 2021-09-17T01:39:56.495.276.762 (XO: 17.495278946)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 129.838 usec -> 2021-09-17T01:39:56.495.276.562 (XO: 17.495278746)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 127.275 usec -> 2021-09-17T01:39:56.495.274.000 (XO: 17.495276184)

 - Block number: 3 starting at 2021-09-17T01:40:20.154.905.725 (XO: 41.154907436 - 41.155213437, t=37.2 degC)  Rank: 00074
   TM values: Max =   161  Min =  -219  Ave =    -4.54  Std =    20.26
   Min TM value:-219 at an offset of 125.188 usec -> 2021-09-17T01:40:20.155.030.912 (XO: 41.155032624)
   Max TM value:161 at an offset of 124.850 usec -> 2021-09-17T01:40:20.155.030.575 (XO: 41.155032287)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 127.538 usec -> 2021-09-17T01:40:20.155.033.262 (XO: 41.155034974)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 124.925 usec -> 2021-09-17T01:40:20.155.030.650 (XO: 41.155032362)

 - Block number: 4 starting at 2021-09-17T01:40:28.143.620.725 (XO: 49.143621899 - 49.143927900, t=37.2 degC)  Rank: 00088
   TM values: Max =   110  Min =  -121  Ave =    -4.97  Std =    28.27
   Min TM value:-121 at an offset of 98.775 usec -> 2021-09-17T01:40:28.143.719.500 (XO: 49.143720674)
   Max TM value:110 at an offset of 145.363 usec -> 2021-09-17T01:40:28.143.766.087 (XO: 49.143767262)
   Max dTM/dt [1/us]:4.26e+02 at an offset of 239.563 usec -> 2021-09-17T01:40:28.143.860.288 (XO: 49.143861462)
   Min dTM/dt [1/us]:-3.19e+02 at an offset of 232.601 usec -> 2021-09-17T01:40:28.143.853.325 (XO: 49.143854500)

 - Block number: 5 starting at 2021-09-17T01:40:43.094.517.725 (XO: 64.094519446 - 64.094825447, t=37.2 degC)  Rank: 00070
   TM values: Max =   155  Min =  -203  Ave =    -4.52  Std =    16.96
   Min TM value:-203 at an offset of 88.438 usec -> 2021-09-17T01:40:43.094.606.162 (XO: 64.094607884)
   Max TM value:155 at an offset of 91.100 usec -> 2021-09-17T01:40:43.094.608.825 (XO: 64.094610547)
   Max dTM/dt [1/us]:1.30e+03 at an offset of 90.913 usec -> 2021-09-17T01:40:43.094.608.637 (XO: 64.094610359)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 88.250 usec -> 2021-09-17T01:40:43.094.605.975 (XO: 64.094607697)
