-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 15:36:20 2022 UTC
-- File 20210916_084017_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.765209 to 54.170285
-each separately from 0s to 54s and from 169996us to 765498us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.989188 in 1 packets: 18
      14.918977 in 1 packets: 36
      22.495755 in 1 packets: 54
MIU packet time from 2021-09-16T08:40:20.765.209.000 to 2021-09-16T08:41:14.170.285.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.989188 in 1 packets: 18
      14.918977 in 1 packets: 36
      22.495755 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9202473 to 55.9200687
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999471 in 1 packets: 18
      14.9999504 in 1 packets: 36
      22.9999239 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210916_084017_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.6854573 to 2.6857463
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9891350  -- OK for SL --
   MIU time from 2021-09-16T08:40:20.765.209.000 to 2021-09-16T08:40:20.765.498.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.989188

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.6748813 to 18.6751703
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9189270  -- OK for SL --
   MIU time from 2021-09-16T08:40:36.754.686.000 to 2021-09-16T08:40:36.754.975.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.918977

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5940973 to 33.5943863
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.4956790  -- OK for SL --
   MIU time from 2021-09-16T08:40:51.673.952.000 to 2021-09-16T08:40:51.674.241.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.495755

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.0900653 to 56.0903543
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-16T08:41:14.169.996.000 to 2021-09-16T08:41:14.170.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.685457300 corrected by 40bit 10MHz counter of 1.920247300 obtained at 1Hz counter of 0. Result: 0.765212525
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-16T08:40:20.765.068.725
   Max 1st sample MIU time among all SL packets: 2021-09-16T08:41:14.170.144.725
   Min 1st sample XO time among all SL packets: 0.765072250
   Max 1st sample XO time among all SL packets: 54.170145886
   TM values from -255 to 215 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-16T08:40:20.765.068.725 (XO: 0.765072250 - 0.765378251, t=37.2 degC)  Rank: 00078
   TM values: Max =   205  Min =  -255  Ave =    -4.56  Std =    20.76
   Min TM value:-255 at an offset of 120.363 usec -> 2021-09-16T08:40:20.765.189.087 (XO: 0.765192613)
   Max TM value:205 at an offset of 120.088 usec -> 2021-09-16T08:40:20.765.188.812 (XO: 0.765192338)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 117.975 usec -> 2021-09-16T08:40:20.765.186.700 (XO: 0.765190225)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 120.200 usec -> 2021-09-16T08:40:20.765.188.925 (XO: 0.765192450)

 - Block number: 2 starting at 2021-09-16T08:40:36.754.545.725 (XO: 16.754549190 - 16.754855192, t=37.2 degC)  Rank: 00082
   TM values: Max =   215  Min =  -243  Ave =    -4.63  Std =    22.11
   Min TM value:-243 at an offset of 120.375 usec -> 2021-09-16T08:40:36.754.666.100 (XO: 16.754669566)
   Max TM value:215 at an offset of 120.075 usec -> 2021-09-16T08:40:36.754.665.800 (XO: 16.754669266)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 122.763 usec -> 2021-09-16T08:40:36.754.668.487 (XO: 16.754671953)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 120.200 usec -> 2021-09-16T08:40:36.754.665.925 (XO: 16.754669391)

 - Block number: 3 starting at 2021-09-16T08:40:51.673.811.725 (XO: 31.673814516 - 31.674120517, t=37.2 degC)  Rank: 00080
   TM values: Max =   197  Min =  -254  Ave =    -4.55  Std =    20.52
   Min TM value:-254 at an offset of 117.713 usec -> 2021-09-16T08:40:51.673.929.437 (XO: 31.673932229)
   Max TM value:197 at an offset of 120.275 usec -> 2021-09-16T08:40:51.673.932.000 (XO: 31.673934791)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 120.125 usec -> 2021-09-16T08:40:51.673.931.850 (XO: 31.673934641)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 117.500 usec -> 2021-09-16T08:40:51.673.929.225 (XO: 31.673932016)

 - Block number: 4 starting at 2021-09-16T08:41:14.169.855.725 (XO: 54.169856886 - 54.170162887, t=37.2 degC)  Rank: 00081
   TM values: Max =   209  Min =  -254  Ave =    -4.50  Std =    24.13
   Min TM value:-254 at an offset of 117.838 usec -> 2021-09-16T08:41:14.169.973.562 (XO: 54.169974723)
   Max TM value:209 at an offset of 117.538 usec -> 2021-09-16T08:41:14.169.973.262 (XO: 54.169974423)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 120.200 usec -> 2021-09-16T08:41:14.169.975.925 (XO: 54.169977086)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 117.638 usec -> 2021-09-16T08:41:14.169.973.362 (XO: 54.169974523)
