-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 15:06:10 2022 UTC
-- File 20210916_081114_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.845633 to 20.835905
-each separately from 6s to 20s and from 835616us to 845922us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.989694 in 1 packets: 18
MIU packet time from 2021-09-16T08:11:23.845.633.000 to 2021-09-16T08:11:37.835.905.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.989694 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8946657 to 21.8946193
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999536 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210916_081114_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7402997 to 8.7405887
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9896470  -- OK for SL --
   MIU time from 2021-09-16T08:11:23.845.633.000 to 2021-09-16T08:11:23.845.922.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.989694

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7302357 to 22.7305247
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-16T08:11:37.835.616.000 to 2021-09-16T08:11:37.835.905.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.740299700 corrected by 40bit 10MHz counter of 7.894665700 obtained at 1Hz counter of 6. Result: 6.845636791
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-16T08:11:23.845.492.725
   Max 1st sample MIU time among all SL packets: 2021-09-16T08:11:37.835.764.725
   Min 1st sample XO time among all SL packets: 6.845496515
   Max 1st sample XO time among all SL packets: 20.835767799
   TM values from -255 to 212 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-16T08:11:23.845.492.725 (XO: 6.845496515 - 6.845802516, t=37.2 degC)  Rank: 00083
   TM values: Max =   189  Min =  -229  Ave =    -4.57  Std =    24.84
   Min TM value:-229 at an offset of 103.000 usec -> 2021-09-16T08:11:23.845.595.725 (XO: 6.845599515)
   Max TM value:189 at an offset of 118.575 usec -> 2021-09-16T08:11:23.845.611.300 (XO: 6.845615091)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 118.450 usec -> 2021-09-16T08:11:23.845.611.175 (XO: 6.845614966)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 161.051 usec -> 2021-09-16T08:11:23.845.653.775 (XO: 6.845657566)

 - Block number: 2 starting at 2021-09-16T08:11:37.835.475.725 (XO: 20.835478799 - 20.835784800, t=37.2 degC)  Rank: 00081
   TM values: Max =   212  Min =  -255  Ave =    -4.59  Std =    20.10
   Min TM value:-255 at an offset of 132.725 usec -> 2021-09-16T08:11:37.835.608.450 (XO: 20.835611524)
   Max TM value:212 at an offset of 135.250 usec -> 2021-09-16T08:11:37.835.610.975 (XO: 20.835614049)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 130.488 usec -> 2021-09-16T08:11:37.835.606.212 (XO: 20.835609286)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 132.600 usec -> 2021-09-16T08:11:37.835.608.325 (XO: 20.835611399)
