-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 07:19:40 2022 UTC
-- File 20210915_234801_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.010813 to 17.878898
-each separately from 6s to 17s and from 10813us to 878898us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.867507 in 1 packets: 18
MIU packet time from 2021-09-15T23:48:10.010.813.000 to 2021-09-15T23:48:21.878.898.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.867507 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8981425 to 18.8981060
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999635 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_234801_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9089562 to 7.9092452
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8674710  -- OK for SL --
   MIU time from 2021-09-15T23:48:10.010.813.000 to 2021-09-15T23:48:10.011.102.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.867507

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.7767162 to 19.7770052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T23:48:21.878.609.000 to 2021-09-15T23:48:21.878.898.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.908956200 corrected by 40bit 10MHz counter of 7.898142500 obtained at 1Hz counter of 6. Result: 6.010813736
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T23:48:10.010.672.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T23:48:21.878.757.725
   Min 1st sample XO time among all SL packets: 6.010673460
   Max 1st sample XO time among all SL packets: 17.878761825
   TM values from -258 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T23:48:10.010.672.725 (XO: 6.010673460 - 6.010979461, t=37.2 degC)  Rank: 00081
   TM values: Max =   232  Min =  -257  Ave =    -4.57  Std =    22.39
   Min TM value:-257 at an offset of 111.000 usec -> 2021-09-15T23:48:10.010.783.725 (XO: 6.010784461)
   Max TM value:232 at an offset of 124.513 usec -> 2021-09-15T23:48:10.010.797.237 (XO: 6.010797973)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 127.125 usec -> 2021-09-15T23:48:10.010.799.850 (XO: 6.010800586)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 124.588 usec -> 2021-09-15T23:48:10.010.797.312 (XO: 6.010798048)

 - Block number: 2 starting at 2021-09-15T23:48:21.878.468.725 (XO: 17.878472824 - 17.878778825, t=37.2 degC)  Rank: 00087
   TM values: Max =   233  Min =  -258  Ave =    -4.46  Std =    28.10
   Min TM value:-258 at an offset of 155.713 usec -> 2021-09-15T23:48:21.878.624.438 (XO: 17.878628537)
   Max TM value:233 at an offset of 116.300 usec -> 2021-09-15T23:48:21.878.585.025 (XO: 17.878589124)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 116.150 usec -> 2021-09-15T23:48:21.878.584.875 (XO: 17.878588974)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 113.600 usec -> 2021-09-15T23:48:21.878.582.325 (XO: 17.878586424)
