-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 11 01:40:18 2022 UTC
-- File 20210914_221642_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.233091 to 26.329478
-each separately from 11s to 26s and from 233091us to 329478us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.095809 in 1 packets: 18
MIU packet time from 2021-09-14T22:16:56.233.091.000 to 2021-09-14T22:17:11.329.478.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.095809 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8575125 to 27.8574628
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999503 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210914_221642_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.0906043 to 13.0908933
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0957590  -- OK for SL --
   MIU time from 2021-09-14T22:16:56.233.091.000 to 2021-09-14T22:16:56.233.380.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.095809

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.1866523 to 28.1869413
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-14T22:17:11.329.189.000 to 2021-09-14T22:17:11.329.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.090604300 corrected by 40bit 10MHz counter of 12.857512500 obtained at 1Hz counter of 11. Result: 11.233092569
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-14T22:16:56.232.950.725
   Max 1st sample MIU time among all SL packets: 2021-09-14T22:17:11.329.337.725
   Min 1st sample XO time among all SL packets: 11.232952294
   Max 1st sample XO time among all SL packets: 26.329339345
   TM values from -268 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-14T22:16:56.232.950.725 (XO: 11.232952294 - 11.233258295, t=37.2 degC)  Rank: 00081
   TM values: Max =   228  Min =  -242  Ave =    -4.58  Std =    19.62
   Min TM value:-242 at an offset of 126.263 usec -> 2021-09-14T22:16:56.233.076.987 (XO: 11.233078557)
   Max TM value:228 at an offset of 128.875 usec -> 2021-09-14T22:16:56.233.079.600 (XO: 11.233081169)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 128.763 usec -> 2021-09-14T22:16:56.233.079.487 (XO: 11.233081057)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 126.063 usec -> 2021-09-14T22:16:56.233.076.787 (XO: 11.233078357)

 - Block number: 2 starting at 2021-09-14T22:17:11.329.048.725 (XO: 26.329050344 - 26.329356345, t=37.2 degC)  Rank: 00084
   TM values: Max =   217  Min =  -268  Ave =    -4.55  Std =    21.80
   Min TM value:-268 at an offset of 123.813 usec -> 2021-09-14T22:17:11.329.172.537 (XO: 26.329174157)
   Max TM value:217 at an offset of 123.538 usec -> 2021-09-14T22:17:11.329.172.262 (XO: 26.329173882)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 126.200 usec -> 2021-09-14T22:17:11.329.174.925 (XO: 26.329176544)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 123.625 usec -> 2021-09-14T22:17:11.329.172.350 (XO: 26.329173969)
