-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 11 01:23:34 2022 UTC
-- File 20210914_220647_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.449267 to 22.625737
-each separately from 1s to 22s and from 449267us to 625737us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.175892 in 1 packets: 18
MIU packet time from 2021-09-14T22:06:51.449.267.000 to 2021-09-14T22:07:12.625.737.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.175892 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9158219 to 23.9157522
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999303 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210914_220647_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3650893 to 3.3653783
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.1758230  -- OK for SL --
   MIU time from 2021-09-14T22:06:51.449.267.000 to 2021-09-14T22:06:51.449.556.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.175892

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5412013 to 24.5414903
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-14T22:07:12.625.448.000 to 2021-09-14T22:07:12.625.737.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.365089300 corrected by 40bit 10MHz counter of 2.915821900 obtained at 1Hz counter of 1. Result: 1.449268883
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-14T22:06:51.449.126.725
   Max 1st sample MIU time among all SL packets: 2021-09-14T22:07:12.625.596.725
   Min 1st sample XO time among all SL packets: 1.449128607
   Max 1st sample XO time among all SL packets: 22.625599889
   TM values from -266 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-14T22:06:51.449.126.725 (XO: 1.449128607 - 1.449434608, t=37.2 degC)  Rank: 00084
   TM values: Max =   254  Min =  -266  Ave =    -4.46  Std =    24.51
   Min TM value:-266 at an offset of 120.138 usec -> 2021-09-14T22:06:51.449.246.862 (XO: 1.449248745)
   Max TM value:254 at an offset of 118.088 usec -> 2021-09-14T22:06:51.449.244.812 (XO: 1.449246695)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 117.913 usec -> 2021-09-14T22:06:51.449.244.637 (XO: 1.449246520)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 120.013 usec -> 2021-09-14T22:06:51.449.246.737 (XO: 1.449248620)

 - Block number: 2 starting at 2021-09-14T22:07:12.625.307.725 (XO: 22.625310889 - 22.625616890, t=37.2 degC)  Rank: 00076
   TM values: Max =   239  Min =  -243  Ave =    -4.60  Std =    27.26
   Min TM value:-243 at an offset of 118.538 usec -> 2021-09-14T22:07:12.625.426.262 (XO: 22.625429426)
   Max TM value:239 at an offset of 121.263 usec -> 2021-09-14T22:07:12.625.428.987 (XO: 22.625432151)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 121.125 usec -> 2021-09-14T22:07:12.625.428.850 (XO: 22.625432014)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 151.088 usec -> 2021-09-14T22:07:12.625.458.812 (XO: 22.625461977)
