-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 10 22:43:45 2022 UTC
-- File 20210914_155844_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.967015 to 24.545837
-each separately from 3s to 24s and from 545548us to 967304us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.578244 in 1 packets: 18
MIU packet time from 2021-09-14T15:58:50.967.015.000 to 2021-09-14T15:59:11.545.837.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.578244 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9080700 to 25.9080003
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999303 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210914_155844_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8750851 to 5.8753741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5781750  -- OK for SL --
   MIU time from 2021-09-14T15:58:50.967.015.000 to 2021-09-14T15:58:50.967.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.578244

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.4535491 to 26.4538381
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-14T15:59:11.545.548.000 to 2021-09-14T15:59:11.545.837.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.875085100 corrected by 40bit 10MHz counter of 4.908070000 obtained at 1Hz counter of 3. Result: 3.967018291
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-14T15:58:50.966.874.725
   Max 1st sample MIU time among all SL packets: 2021-09-14T15:59:11.545.696.725
   Min 1st sample XO time among all SL packets: 3.966878016
   Max 1st sample XO time among all SL packets: 24.545699326
   TM values from -261 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-14T15:58:50.966.874.725 (XO: 3.966878016 - 3.967184017, t=37.2 degC)  Rank: 00079
   TM values: Max =   233  Min =  -261  Ave =    -4.51  Std =    21.49
   Min TM value:-261 at an offset of 91.713 usec -> 2021-09-14T15:58:50.966.966.437 (XO: 3.966969729)
   Max TM value:233 at an offset of 91.450 usec -> 2021-09-14T15:58:50.966.966.175 (XO: 3.966969466)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 94.100 usec -> 2021-09-14T15:58:50.966.968.825 (XO: 3.966972116)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 91.563 usec -> 2021-09-14T15:58:50.966.966.287 (XO: 3.966969578)

 - Block number: 2 starting at 2021-09-14T15:59:11.545.407.725 (XO: 24.545410325 - 24.545716326, t=37.2 degC)  Rank: 00087
   TM values: Max =   206  Min =  -232  Ave =    -4.49  Std =    23.20
   Min TM value:-232 at an offset of 126.588 usec -> 2021-09-14T15:59:11.545.534.312 (XO: 24.545536913)
   Max TM value:206 at an offset of 129.300 usec -> 2021-09-14T15:59:11.545.537.025 (XO: 24.545539625)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 129.125 usec -> 2021-09-14T15:59:11.545.536.850 (XO: 24.545539450)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 126.388 usec -> 2021-09-14T15:59:11.545.534.112 (XO: 24.545536713)
