-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  9 07:37:38 2022 UTC
-- File 20210911_150209_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.965965 to 26.498336
-each separately from 6s to 26s and from 498047us to 966254us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.531793 in 1 packets: 18
MIU packet time from 2021-09-11T15:02:18.965.965.000 to 2021-09-11T15:02:38.498.336.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.531793 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8876415 to 27.8875747
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999332 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210911_150209_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8536069 to 8.8538959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5317270  -- OK for SL --
   MIU time from 2021-09-11T15:02:18.965.965.000 to 2021-09-11T15:02:18.966.254.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.531793

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3856229 to 28.3859119
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-11T15:02:38.498.047.000 to 2021-09-11T15:02:38.498.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.853606900 corrected by 40bit 10MHz counter of 7.887641500 obtained at 1Hz counter of 6. Result: 6.965968588
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-11T15:02:18.965.824.725
   Max 1st sample MIU time among all SL packets: 2021-09-11T15:02:38.498.195.725
   Min 1st sample XO time among all SL packets: 6.965828312
   Max 1st sample XO time among all SL packets: 26.498198619
   TM values from -248 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-11T15:02:18.965.824.725 (XO: 6.965828312 - 6.966134313, t=35.6 degC)  Rank: 00078
   TM values: Max =   222  Min =  -232  Ave =    -4.56  Std =    18.38
   Min TM value:-232 at an offset of 124.275 usec -> 2021-09-11T15:02:18.965.949.000 (XO: 6.965952588)
   Max TM value:222 at an offset of 126.963 usec -> 2021-09-11T15:02:18.965.951.687 (XO: 6.965955275)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 126.775 usec -> 2021-09-11T15:02:18.965.951.500 (XO: 6.965955088)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 124.050 usec -> 2021-09-11T15:02:18.965.948.775 (XO: 6.965952363)

 - Block number: 2 starting at 2021-09-11T15:02:38.497.906.725 (XO: 26.497909618 - 26.498215619, t=35.6 degC)  Rank: 00082
   TM values: Max =   201  Min =  -248  Ave =    -4.48  Std =    23.81
   Min TM value:-248 at an offset of 230.251 usec -> 2021-09-11T15:02:38.498.136.975 (XO: 26.498139869)
   Max TM value:201 at an offset of 126.600 usec -> 2021-09-11T15:02:38.498.033.325 (XO: 26.498036218)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 126.450 usec -> 2021-09-11T15:02:38.498.033.175 (XO: 26.498036068)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 230.163 usec -> 2021-09-11T15:02:38.498.136.888 (XO: 26.498139781)
