-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan  6 09:10:09 2022 UTC
-- File 20210907_233213_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.026694 to 73.847279
-each separately from 3s to 73s and from 26694us to 847279us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.199809 in 1 packets: 18
      12.599799 in 1 packets: 36
      13.939774 in 1 packets: 54
      24.079758 in 1 packets: 72
MIU packet time from 2021-09-07T23:32:19.026.694.000 to 2021-09-07T23:33:29.847.279.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.199809 in 1 packets: 18
      12.599799 in 1 packets: 36
      13.939774 in 1 packets: 54
      24.079758 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9237347 to 74.9235028
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999338 in 1 packets: 18
      11.9999602 in 1 packets: 36
      13.9999536 in 1 packets: 54
      23.9999205 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210907_233213_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.9504296 to 4.9507186
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1997430  -- OK for SL --
   MIU time from 2021-09-07T23:32:19.026.694.000 to 2021-09-07T23:32:19.026.983.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.199809

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1504616 to 25.1507506
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5997590  -- OK for SL --
   MIU time from 2021-09-07T23:32:39.226.792.000 to 2021-09-07T23:32:39.227.081.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.599799

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.7505096 to 37.7507986
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9397270  -- OK for SL --
   MIU time from 2021-09-07T23:32:51.826.880.000 to 2021-09-07T23:32:51.827.169.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.939774

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.6905256 to 51.6908146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0796790  -- OK for SL --
   MIU time from 2021-09-07T23:33:05.766.943.000 to 2021-09-07T23:33:05.767.232.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.079758

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.7704936 to 75.7707826
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-07T23:33:29.846.990.000 to 2021-09-07T23:33:29.847.279.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.950429600 corrected by 40bit 10MHz counter of 4.923734700 obtained at 1Hz counter of 3. Result: 3.026694988
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-07T23:32:19.026.553.725
   Max 1st sample MIU time among all SL packets: 2021-09-07T23:33:29.847.138.725
   Min 1st sample XO time among all SL packets: 3.026554713
   Max 1st sample XO time among all SL packets: 73.847142321
   TM values from -243 to 214 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-07T23:32:19.026.553.725 (XO: 3.026554713 - 3.026860714, t=37.2 degC)  Rank: 00079
   TM values: Max =   204  Min =  -243  Ave =    -4.38  Std =    18.29
   Min TM value:-243 at an offset of 109.688 usec -> 2021-09-07T23:32:19.026.663.412 (XO: 3.026664400)
   Max TM value:204 at an offset of 112.325 usec -> 2021-09-07T23:32:19.026.666.050 (XO: 3.026667038)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 112.175 usec -> 2021-09-07T23:32:19.026.665.900 (XO: 3.026666888)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 109.475 usec -> 2021-09-07T23:32:19.026.663.200 (XO: 3.026664188)

 - Block number: 2 starting at 2021-09-07T23:32:39.226.651.725 (XO: 23.226653596 - 23.226959597, t=37.2 degC)  Rank: 00078
   TM values: Max =   214  Min =  -219  Ave =    -4.43  Std =    20.49
   Min TM value:-219 at an offset of 109.750 usec -> 2021-09-07T23:32:39.226.761.475 (XO: 23.226763346)
   Max TM value:214 at an offset of 112.413 usec -> 2021-09-07T23:32:39.226.764.137 (XO: 23.226766009)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 112.250 usec -> 2021-09-07T23:32:39.226.763.975 (XO: 23.226765846)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 109.525 usec -> 2021-09-07T23:32:39.226.761.250 (XO: 23.226763121)

 - Block number: 3 starting at 2021-09-07T23:32:51.826.739.725 (XO: 35.826743353 - 35.827049354, t=37.2 degC)  Rank: 00081
   TM values: Max =   214  Min =  -218  Ave =    -4.51  Std =    20.09
   Min TM value:-218 at an offset of 109.050 usec -> 2021-09-07T23:32:51.826.848.775 (XO: 35.826852404)
   Max TM value:214 at an offset of 111.738 usec -> 2021-09-07T23:32:51.826.851.462 (XO: 35.826855091)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 111.613 usec -> 2021-09-07T23:32:51.826.851.337 (XO: 35.826854966)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 108.900 usec -> 2021-09-07T23:32:51.826.848.625 (XO: 35.826852254)

 - Block number: 4 starting at 2021-09-07T23:33:05.766.802.725 (XO: 49.766805555 - 49.767111556, t=37.2 degC)  Rank: 00078
   TM values: Max =   208  Min =  -221  Ave =    -4.49  Std =    19.90
   Min TM value:-221 at an offset of 107.863 usec -> 2021-09-07T23:33:05.766.910.587 (XO: 49.766913418)
   Max TM value:208 at an offset of 123.338 usec -> 2021-09-07T23:33:05.766.926.062 (XO: 49.766928893)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 123.213 usec -> 2021-09-07T23:33:05.766.925.937 (XO: 49.766928768)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 120.463 usec -> 2021-09-07T23:33:05.766.923.187 (XO: 49.766926018)

 - Block number: 5 starting at 2021-09-07T23:33:29.846.849.725 (XO: 73.846853320 - 73.847159321, t=37.2 degC)  Rank: 00076
   TM values: Max =   214  Min =  -241  Ave =    -4.44  Std =    20.74
   Min TM value:-241 at an offset of 125.125 usec -> 2021-09-07T23:33:29.846.974.850 (XO: 73.846978445)
   Max TM value:214 at an offset of 127.763 usec -> 2021-09-07T23:33:29.846.977.487 (XO: 73.846981083)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 127.575 usec -> 2021-09-07T23:33:29.846.977.300 (XO: 73.846980895)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 112.188 usec -> 2021-09-07T23:33:29.846.961.912 (XO: 73.846965507)
