-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan  6 05:46:24 2022 UTC
-- File 20210907_184829_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.706395 to 93.050031
-each separately from 1s to 93s and from 49742us to 837385us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.130412 in 1 packets: 18
      8.710157 in 1 packets: 36
      20.110386 in 1 packets: 54
      14.460192 in 1 packets: 72
      21.110614 in 1 packets: 90
      3.819852 in 1 packets: 108
MIU packet time from 2021-09-07T18:48:33.706.395.000 to 2021-09-07T18:50:05.050.031.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.130412 in 1 packets: 18
      8.710157 in 1 packets: 36
      20.110386 in 1 packets: 54
      14.460192 in 1 packets: 72
      21.110614 in 1 packets: 90
      3.819852 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.6199795 to 94.6196722
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999231 in 1 packets: 18
      8.9999699 in 1 packets: 36
      19.9999333 in 1 packets: 54
      14.9999499 in 1 packets: 72
      20.9999298 in 1 packets: 90
      3.9999867 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210907_184829_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3263751 to 3.3266641
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1303350  -- OK for SL --
   MIU time from 2021-09-07T18:48:33.706.395.000 to 2021-09-07T18:48:33.706.684.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.130412

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.4569991 to 26.4572881
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7101270  -- OK for SL --
   MIU time from 2021-09-07T18:48:56.837.096.000 to 2021-09-07T18:48:56.837.385.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.710157

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.1674151 to 35.1677041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1103190  -- OK for SL --
   MIU time from 2021-09-07T18:49:05.547.542.000 to 2021-09-07T18:49:05.547.831.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.110386

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.2780231 to 55.2783121
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4601430  -- OK for SL --
   MIU time from 2021-09-07T18:49:25.658.217.000 to 2021-09-07T18:49:25.658.506.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.460192

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.7384551 to 69.7387441
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.1105430  -- OK for SL --
   MIU time from 2021-09-07T18:49:40.118.698.000 to 2021-09-07T18:49:40.118.987.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.110614

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 90.8492871 to 90.8495761
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.8198390  -- OK for SL --
   MIU time from 2021-09-07T18:50:01.229.601.000 to 2021-09-07T18:50:01.229.890.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.819852

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 94.6694151 to 94.6697041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-07T18:50:05.049.742.000 to 2021-09-07T18:50:05.050.031.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.326375100 corrected by 40bit 10MHz counter of 2.619979500 obtained at 1Hz counter of 1. Result: 1.706398002
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-07T18:48:33.706.254.725
   Max 1st sample MIU time among all SL packets: 2021-09-07T18:50:05.049.890.725
   Min 1st sample XO time among all SL packets: 1.706257726
   Max 1st sample XO time among all SL packets: 93.049891790
   TM values from -315 to 320 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-07T18:48:33.706.254.725 (XO: 1.706257726 - 1.706563727, t=37.2 degC)  Rank: 00072
   TM values: Max =   320  Min =  -315  Ave =    -4.41  Std =    41.72
   Min TM value:-315 at an offset of 120.938 usec -> 2021-09-07T18:48:33.706.375.662 (XO: 1.706378664)
   Max TM value:320 at an offset of 123.563 usec -> 2021-09-07T18:48:33.706.378.287 (XO: 1.706381289)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 123.375 usec -> 2021-09-07T18:48:33.706.378.100 (XO: 1.706381102)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 120.763 usec -> 2021-09-07T18:48:33.706.375.487 (XO: 1.706378489)

 - Block number: 2 starting at 2021-09-07T18:48:56.836.955.725 (XO: 24.836958987 - 24.837264988, t=37.2 degC)  Rank: 00079
   TM values: Max =   229  Min =  -298  Ave =    -4.54  Std =    26.63
   Min TM value:-298 at an offset of 110.388 usec -> 2021-09-07T18:48:56.837.066.112 (XO: 24.837069375)
   Max TM value:229 at an offset of 112.925 usec -> 2021-09-07T18:48:56.837.068.650 (XO: 24.837071912)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 107.975 usec -> 2021-09-07T18:48:56.837.063.700 (XO: 24.837066962)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 98.550 usec -> 2021-09-07T18:48:56.837.054.275 (XO: 24.837057537)

 - Block number: 3 starting at 2021-09-07T18:49:05.547.401.725 (XO: 33.547404131 - 33.547710132, t=37.2 degC)  Rank: 00074
   TM values: Max =   250  Min =  -300  Ave =    -4.42  Std =    31.63
   Min TM value:-300 at an offset of 95.788 usec -> 2021-09-07T18:49:05.547.497.512 (XO: 33.547499919)
   Max TM value:250 at an offset of 98.338 usec -> 2021-09-07T18:49:05.547.500.062 (XO: 33.547502469)
   Max dTM/dt [1/us]:2.89e+03 at an offset of 98.163 usec -> 2021-09-07T18:49:05.547.499.887 (XO: 33.547502294)
   Min dTM/dt [1/us]:-2.65e+03 at an offset of 95.650 usec -> 2021-09-07T18:49:05.547.497.375 (XO: 33.547499782)

 - Block number: 4 starting at 2021-09-07T18:49:25.658.076.725 (XO: 53.658079197 - 53.658385198, t=37.2 degC)  Rank: 00086
   TM values: Max =   251  Min =  -285  Ave =    -4.52  Std =    27.00
   Min TM value:-285 at an offset of 107.925 usec -> 2021-09-07T18:49:25.658.184.650 (XO: 53.658187122)
   Max TM value:251 at an offset of 123.650 usec -> 2021-09-07T18:49:25.658.200.375 (XO: 53.658202847)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 123.463 usec -> 2021-09-07T18:49:25.658.200.187 (XO: 53.658202660)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 120.925 usec -> 2021-09-07T18:49:25.658.197.650 (XO: 53.658200122)

 - Block number: 5 starting at 2021-09-07T18:49:40.118.557.725 (XO: 68.118559528 - 68.118865529, t=37.2 degC)  Rank: 00075
   TM values: Max =   250  Min =  -284  Ave =    -4.41  Std =    34.82
   Min TM value:-284 at an offset of 112.450 usec -> 2021-09-07T18:49:40.118.670.175 (XO: 68.118671978)
   Max TM value:250 at an offset of 115.013 usec -> 2021-09-07T18:49:40.118.672.737 (XO: 68.118674541)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 114.825 usec -> 2021-09-07T18:49:40.118.672.550 (XO: 68.118674353)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 112.300 usec -> 2021-09-07T18:49:40.118.670.025 (XO: 68.118671828)

 - Block number: 6 starting at 2021-09-07T18:50:01.229.460.725 (XO: 89.229462082 - 89.229768083, t=37.2 degC)  Rank: 00079
   TM values: Max =   252  Min =  -286  Ave =    -4.44  Std =    26.72
   Min TM value:-286 at an offset of 120.638 usec -> 2021-09-07T18:50:01.229.581.362 (XO: 89.229582720)
   Max TM value:252 at an offset of 123.238 usec -> 2021-09-07T18:50:01.229.583.962 (XO: 89.229585320)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 123.063 usec -> 2021-09-07T18:50:01.229.583.787 (XO: 89.229585145)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 108.600 usec -> 2021-09-07T18:50:01.229.569.325 (XO: 89.229570683)

 - Block number: 7 starting at 2021-09-07T18:50:05.049.601.725 (XO: 93.049602789 - 93.049908790, t=37.2 degC)  Rank: 00081
   TM values: Max =   255  Min =  -257  Ave =    -4.38  Std =    28.97
   Min TM value:-257 at an offset of 107.388 usec -> 2021-09-07T18:50:05.049.709.112 (XO: 93.049710177)
   Max TM value:255 at an offset of 122.525 usec -> 2021-09-07T18:50:05.049.724.250 (XO: 93.049725314)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 122.388 usec -> 2021-09-07T18:50:05.049.724.112 (XO: 93.049725177)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 119.788 usec -> 2021-09-07T18:50:05.049.721.512 (XO: 93.049722577)
