-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan  6 02:03:41 2022 UTC
-- File 20210907_150604_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.958461 to 28.210653
-each separately from 4s to 28s and from 210364us to 958750us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.251614 in 1 packets: 18
MIU packet time from 2021-09-07T15:06:11.958.461.000 to 2021-09-07T15:06:35.210.653.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.251614 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7399068 to 29.7398268
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999200 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210907_150604_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.6983679 to 6.6986569
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.2515350  -- OK for SL --
   MIU time from 2021-09-07T15:06:11.958.461.000 to 2021-09-07T15:06:11.958.750.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.251614

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9501919 to 29.9504809
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-07T15:06:35.210.364.000 to 2021-09-07T15:06:35.210.653.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.698367900 corrected by 40bit 10MHz counter of 5.739906800 obtained at 1Hz counter of 4. Result: 4.958464263
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-07T15:06:11.958.320.725
   Max 1st sample MIU time among all SL packets: 2021-09-07T15:06:35.210.512.725
   Min 1st sample XO time among all SL packets: 4.958323987
   Max 1st sample XO time among all SL packets: 28.210514520
   TM values from -278 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-07T15:06:11.958.320.725 (XO: 4.958323987 - 4.958629988, t=35.6 degC)  Rank: 00080
   TM values: Max =   236  Min =  -260  Ave =    -4.46  Std =    20.16
   Min TM value:-260 at an offset of 127.250 usec -> 2021-09-07T15:06:11.958.447.975 (XO: 4.958451238)
   Max TM value:236 at an offset of 129.888 usec -> 2021-09-07T15:06:11.958.450.612 (XO: 4.958453875)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 129.738 usec -> 2021-09-07T15:06:11.958.450.462 (XO: 4.958453725)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 127.063 usec -> 2021-09-07T15:06:11.958.447.787 (XO: 4.958451050)

 - Block number: 2 starting at 2021-09-07T15:06:35.210.223.725 (XO: 28.210225519 - 28.210531520, t=35.6 degC)  Rank: 00082
   TM values: Max =   218  Min =  -278  Ave =    -4.61  Std =    22.09
   Min TM value:-278 at an offset of 132.813 usec -> 2021-09-07T15:06:35.210.356.537 (XO: 28.210358332)
   Max TM value:218 at an offset of 135.438 usec -> 2021-09-07T15:06:35.210.359.162 (XO: 28.210360957)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 135.225 usec -> 2021-09-07T15:06:35.210.358.950 (XO: 28.210360744)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 132.625 usec -> 2021-09-07T15:06:35.210.356.350 (XO: 28.210358144)
