-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 18:13:52 2022 UTC
-- File 20210904_085351_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.831438 to 26.039697
-each separately from 1s to 26s and from 39408us to 831727us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.207681 in 1 packets: 18
MIU packet time from 2021-09-04T08:53:55.831.438.000 to 2021-09-04T08:54:20.039.697.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.207681 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8715815 to 27.8714985
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999170 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210904_085351_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7030199 to 3.7033089
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.2075990  -- OK for SL --
   MIU time from 2021-09-04T08:53:55.831.438.000 to 2021-09-04T08:53:55.831.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.207681

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.9109079 to 27.9111969
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-04T08:54:20.039.408.000 to 2021-09-04T08:54:20.039.697.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.703019900 corrected by 40bit 10MHz counter of 2.871581500 obtained at 1Hz counter of 1. Result: 1.831441144
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-04T08:53:55.831.297.725
   Max 1st sample MIU time among all SL packets: 2021-09-04T08:54:20.039.556.725
   Min 1st sample XO time among all SL packets: 1.831300868
   Max 1st sample XO time among all SL packets: 26.039558259
   TM values from -221 to 182 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-04T08:53:55.831.297.725 (XO: 1.831300868 - 1.831606869, t=37.2 degC)  Rank: 00074
   TM values: Max =   175  Min =  -221  Ave =    -4.54  Std =    17.64
   Min TM value:-221 at an offset of 109.675 usec -> 2021-09-04T08:53:55.831.407.400 (XO: 1.831410544)
   Max TM value:175 at an offset of 112.250 usec -> 2021-09-04T08:53:55.831.409.975 (XO: 1.831413119)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 112.113 usec -> 2021-09-04T08:53:55.831.409.837 (XO: 1.831412981)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 109.488 usec -> 2021-09-04T08:53:55.831.407.212 (XO: 1.831410356)

 - Block number: 2 starting at 2021-09-04T08:54:20.039.267.725 (XO: 26.039269259 - 26.039575260, t=37.2 degC)  Rank: 00086
   TM values: Max =   182  Min =  -216  Ave =    -4.40  Std =    18.71
   Min TM value:-216 at an offset of 107.413 usec -> 2021-09-04T08:54:20.039.375.137 (XO: 26.039376671)
   Max TM value:182 at an offset of 109.950 usec -> 2021-09-04T08:54:20.039.377.675 (XO: 26.039379209)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 109.800 usec -> 2021-09-04T08:54:20.039.377.525 (XO: 26.039379059)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 107.238 usec -> 2021-09-04T08:54:20.039.374.962 (XO: 26.039376496)
