-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan  2 15:14:13 2022 UTC
-- File 20210903_160540_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.687594 to 32.654024
-each separately from 4s to 32s and from 653735us to 687883us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.983870 in 1 packets: 18
      8.981693 in 1 packets: 36
MIU packet time from 2021-09-03T16:05:47.687.594.000 to 2021-09-03T16:06:15.654.024.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.983870 in 1 packets: 18
      8.981693 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8950844 to 33.8949909
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999366 in 1 packets: 18
      8.9999699 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210903_160540_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5826791 to 6.5829681
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9838070  -- OK for SL --
   MIU time from 2021-09-03T16:05:47.687.594.000 to 2021-09-03T16:05:47.687.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.983870

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.5667751 to 25.5670641
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9816630  -- OK for SL --
   MIU time from 2021-09-03T16:06:06.671.753.000 to 2021-09-03T16:06:06.672.042.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.981693

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.5487271 to 34.5490161
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-03T16:06:15.653.735.000 to 2021-09-03T16:06:15.654.024.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.582679100 corrected by 40bit 10MHz counter of 5.895084400 obtained at 1Hz counter of 4. Result: 4.687596969
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-03T16:05:47.687.453.725
   Max 1st sample MIU time among all SL packets: 2021-09-03T16:06:15.653.883.725
   Min 1st sample XO time among all SL packets: 4.687456694
   Max 1st sample XO time among all SL packets: 32.653887083
   TM values from -283 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-03T16:05:47.687.453.725 (XO: 4.687456694 - 4.687762695, t=37.2 degC)  Rank: 00079
   TM values: Max =   227  Min =  -258  Ave =    -4.49  Std =    23.23
   Min TM value:-258 at an offset of 128.050 usec -> 2021-09-03T16:05:47.687.581.775 (XO: 4.687584744)
   Max TM value:227 at an offset of 128.325 usec -> 2021-09-03T16:05:47.687.582.050 (XO: 4.687585019)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 128.150 usec -> 2021-09-03T16:05:47.687.581.875 (XO: 4.687584844)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 125.625 usec -> 2021-09-03T16:05:47.687.579.350 (XO: 4.687582319)

 - Block number: 2 starting at 2021-09-03T16:06:06.671.612.725 (XO: 23.671616041 - 23.671922042, t=37.2 degC)  Rank: 00086
   TM values: Max =   216  Min =  -251  Ave =    -4.45  Std =    20.71
   Min TM value:-251 at an offset of 124.775 usec -> 2021-09-03T16:06:06.671.737.500 (XO: 23.671740817)
   Max TM value:216 at an offset of 127.325 usec -> 2021-09-03T16:06:06.671.740.050 (XO: 23.671743367)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 127.188 usec -> 2021-09-03T16:06:06.671.739.912 (XO: 23.671743229)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 124.600 usec -> 2021-09-03T16:06:06.671.737.325 (XO: 23.671740642)

 - Block number: 3 starting at 2021-09-03T16:06:15.653.594.725 (XO: 32.653598082 - 32.653904083, t=37.2 degC)  Rank: 00088
   TM values: Max =   238  Min =  -283  Ave =    -4.54  Std =    24.33
   Min TM value:-283 at an offset of 127.613 usec -> 2021-09-03T16:06:15.653.722.337 (XO: 32.653725695)
   Max TM value:238 at an offset of 130.150 usec -> 2021-09-03T16:06:15.653.724.875 (XO: 32.653728232)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 130.025 usec -> 2021-09-03T16:06:15.653.724.750 (XO: 32.653728107)
   Min dTM/dt [1/us]:-2.72e+03 at an offset of 127.488 usec -> 2021-09-03T16:06:15.653.722.212 (XO: 32.653725570)
