-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  1 16:42:11 2022 UTC
-- File 20210902_232441_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.778051 to 27.536391
-each separately from 2s to 27s and from 536102us to 778340us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.757762 in 1 packets: 18
MIU packet time from 2021-09-02T23:24:46.778.051.000 to 2021-09-02T23:25:11.536.391.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.757762 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8998767 to 28.8997938
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999171 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210902_232441_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6779286 to 4.6782176
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.7576790  -- OK for SL --
   MIU time from 2021-09-02T23:24:46.778.051.000 to 2021-09-02T23:24:46.778.340.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.757762

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.4358966 to 29.4361856
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-02T23:25:11.536.102.000 to 2021-09-02T23:25:11.536.391.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.677928600 corrected by 40bit 10MHz counter of 3.899876700 obtained at 1Hz counter of 2. Result: 2.778054468
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-02T23:24:46.777.910.725
   Max 1st sample MIU time among all SL packets: 2021-09-02T23:25:11.536.250.725
   Min 1st sample XO time among all SL packets: 2.777914192
   Max 1st sample XO time among all SL packets: 27.536253295
   TM values from -246 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-02T23:24:46.777.910.725 (XO: 2.777914192 - 2.778220193, t=37.2 degC)  Rank: 00083
   TM values: Max =   215  Min =  -235  Ave =    -4.38  Std =    19.59
   Min TM value:-235 at an offset of 127.575 usec -> 2021-09-02T23:24:46.778.038.300 (XO: 2.778041768)
   Max TM value:215 at an offset of 130.238 usec -> 2021-09-02T23:24:46.778.040.962 (XO: 2.778044430)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 130.038 usec -> 2021-09-02T23:24:46.778.040.762 (XO: 2.778044230)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 127.438 usec -> 2021-09-02T23:24:46.778.038.162 (XO: 2.778041630)

 - Block number: 2 starting at 2021-09-02T23:25:11.535.961.725 (XO: 27.535964294 - 27.536270295, t=37.2 degC)  Rank: 00091
   TM values: Max =   208  Min =  -246  Ave =    -4.48  Std =    20.50
   Min TM value:-246 at an offset of 101.738 usec -> 2021-09-02T23:25:11.536.063.462 (XO: 27.536066032)
   Max TM value:208 at an offset of 116.925 usec -> 2021-09-02T23:25:11.536.078.650 (XO: 27.536081219)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 116.750 usec -> 2021-09-02T23:25:11.536.078.475 (XO: 27.536081044)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 114.100 usec -> 2021-09-02T23:25:11.536.075.825 (XO: 27.536078394)
