-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  1 16:00:43 2022 UTC
-- File 20210902_224808_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.244250 to 75.580993
-each separately from 1s to 75s and from 81029us to 863572us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.618744 in 1 packets: 18
      18.858782 in 1 packets: 36
      8.039402 in 1 packets: 54
      21.318695 in 1 packets: 72
      8.499386 in 1 packets: 90
MIU packet time from 2021-09-02T22:48:12.244.250.000 to 2021-09-02T22:49:26.580.993.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.618744 in 1 packets: 18
      18.858782 in 1 packets: 36
      8.039402 in 1 packets: 54
      21.318695 in 1 packets: 72
      8.499386 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8474941 to 76.8472485
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999435 in 1 packets: 18
      18.9999370 in 1 packets: 36
      7.9999734 in 2 packets: 54, 90
      21.9999271 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      06 -> 47.2 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210902_224808_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0917455 to 3.0920345
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6186870  -- OK for SL --
   MIU time from 2021-09-02T22:48:12.244.250.000 to 2021-09-02T22:48:12.244.539.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.618744

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7107215 to 20.7110105
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8587190  -- OK for SL --
   MIU time from 2021-09-02T22:48:29.863.283.000 to 2021-09-02T22:48:29.863.572.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.858782

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.5697295 to 39.5700185
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0393750  -- OK for SL --
   MIU time from 2021-09-02T22:48:48.722.354.000 to 2021-09-02T22:48:48.722.643.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.039402

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.6093935 to 47.6096825
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3186230  -- OK for SL --
   MIU time from 2021-09-02T22:48:56.762.045.000 to 2021-09-02T22:48:56.762.334.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.318695

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.9283055 to 68.9285945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4993590  -- OK for SL --
   MIU time from 2021-09-02T22:49:18.081.029.000 to 2021-09-02T22:49:18.081.318.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.499386

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.4279535 to 77.4282425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-02T22:49:26.580.704.000 to 2021-09-02T22:49:26.580.993.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.091745500 corrected by 40bit 10MHz counter of 2.847494100 obtained at 1Hz counter of 1. Result: 1.244252206
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-02T22:48:12.244.109.725
   Max 1st sample MIU time among all SL packets: 2021-09-02T22:49:26.580.852.725
   Min 1st sample XO time among all SL packets: 1.244111931
   Max 1st sample XO time among all SL packets: 75.580855642
   TM values from -315 to 247 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-02T22:48:12.244.109.725 (XO: 1.244111931 - 1.244417932, t=37.2 degC)  Rank: 00082
   TM values: Max =   247  Min =  -315  Ave =    -4.48  Std =    23.85
   Min TM value:-315 at an offset of 126.800 usec -> 2021-09-02T22:48:12.244.236.525 (XO: 1.244238731)
   Max TM value:247 at an offset of 126.550 usec -> 2021-09-02T22:48:12.244.236.275 (XO: 1.244238481)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 124.588 usec -> 2021-09-02T22:48:12.244.234.312 (XO: 1.244236518)
   Min dTM/dt [1/us]:-2.74e+03 at an offset of 126.675 usec -> 2021-09-02T22:48:12.244.236.400 (XO: 1.244238606)

 - Block number: 2 starting at 2021-09-02T22:48:29.863.142.725 (XO: 18.863146473 - 18.863452474, t=37.2 degC)  Rank: 00080
   TM values: Max =   226  Min =  -263  Ave =    -4.44  Std =    22.23
   Min TM value:-263 at an offset of 127.138 usec -> 2021-09-02T22:48:29.863.269.862 (XO: 18.863273611)
   Max TM value:226 at an offset of 126.875 usec -> 2021-09-02T22:48:29.863.269.600 (XO: 18.863273349)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 129.538 usec -> 2021-09-02T22:48:29.863.272.262 (XO: 18.863276011)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 127.000 usec -> 2021-09-02T22:48:29.863.269.725 (XO: 18.863273474)

 - Block number: 3 starting at 2021-09-02T22:48:48.722.213.725 (XO: 37.722217008 - 37.722523009, t=37.2 degC)  Rank: 00079
   TM values: Max =   231  Min =  -231  Ave =    -4.40  Std =    18.86
   Min TM value:-231 at an offset of 126.338 usec -> 2021-09-02T22:48:48.722.340.062 (XO: 37.722343346)
   Max TM value:231 at an offset of 129.038 usec -> 2021-09-02T22:48:48.722.342.762 (XO: 37.722346046)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 128.888 usec -> 2021-09-02T22:48:48.722.342.612 (XO: 37.722345896)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 126.138 usec -> 2021-09-02T22:48:48.722.339.862 (XO: 37.722343146)

 - Block number: 4 starting at 2021-09-02T22:48:56.761.904.725 (XO: 45.761907739 - 45.762213740, t=37.2 degC)  Rank: 00078
   TM values: Max =   218  Min =  -261  Ave =    -4.58  Std =    31.82
   Min TM value:-261 at an offset of 158.151 usec -> 2021-09-02T22:48:56.762.062.875 (XO: 45.762065890)
   Max TM value:218 at an offset of 197.126 usec -> 2021-09-02T22:48:56.762.101.850 (XO: 45.762104865)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 130.788 usec -> 2021-09-02T22:48:56.762.035.512 (XO: 45.762038527)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 128.213 usec -> 2021-09-02T22:48:56.762.032.937 (XO: 45.762035952)

 - Block number: 5 starting at 2021-09-02T22:49:18.080.888.725 (XO: 67.080890392 - 67.081196393, t=37.2 degC)  Rank: 00080
   TM values: Max =   226  Min =  -263  Ave =    -4.51  Std =    22.68
   Min TM value:-263 at an offset of 112.913 usec -> 2021-09-02T22:49:18.081.001.637 (XO: 67.081003305)
   Max TM value:226 at an offset of 115.450 usec -> 2021-09-02T22:49:18.081.004.175 (XO: 67.081005842)
   Max dTM/dt [1/us]:2.38e+03 at an offset of 115.288 usec -> 2021-09-02T22:49:18.081.004.012 (XO: 67.081005680)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 112.738 usec -> 2021-09-02T22:49:18.081.001.462 (XO: 67.081003130)

 - Block number: 6 starting at 2021-09-02T22:49:26.580.563.725 (XO: 75.580566641 - 75.580872642, t=37.2 degC)  Rank: 00080
   TM values: Max =   207  Min =  -249  Ave =    -4.41  Std =    24.43
   Min TM value:-249 at an offset of 112.550 usec -> 2021-09-02T22:49:26.580.676.275 (XO: 75.580679191)
   Max TM value:207 at an offset of 127.725 usec -> 2021-09-02T22:49:26.580.691.450 (XO: 75.580694366)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 127.563 usec -> 2021-09-02T22:49:26.580.691.287 (XO: 75.580694204)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 124.863 usec -> 2021-09-02T22:49:26.580.688.587 (XO: 75.580691504)
