-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan  1 13:47:30 2022 UTC
-- File 20210902_203639_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.103151 to 47.986808
-each separately from 11s to 47s and from 4227us to 986808us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.900787 in 1 packets: 18
      11.840772 in 1 packets: 36
      14.140942 in 1 packets: 54
MIU packet time from 2021-09-02T20:36:53.103.151.000 to 2021-09-02T20:37:29.986.808.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.900787 in 1 packets: 18
      11.840772 in 1 packets: 36
      14.140942 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8246633 to 48.8245438
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999635 in 1 packets: 18
      10.9999634 in 1 packets: 36
      13.9999536 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210902_203639_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9278152 to 12.9281042
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9007510  -- OK for SL --
   MIU time from 2021-09-02T20:36:53.103.151.000 to 2021-09-02T20:36:53.103.440.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.900787

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8288552 to 23.8291442
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8407350  -- OK for SL --
   MIU time from 2021-09-02T20:37:04.004.227.000 to 2021-09-02T20:37:04.004.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.840772

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6698792 to 35.6701682
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1408950  -- OK for SL --
   MIU time from 2021-09-02T20:37:15.845.288.000 to 2021-09-02T20:37:15.845.577.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.140942

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.8110632 to 49.8113522
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-02T20:37:29.986.519.000 to 2021-09-02T20:37:29.986.808.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.927815200 corrected by 40bit 10MHz counter of 12.824663300 obtained at 1Hz counter of 11. Result: 11.103152240
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-02T20:36:53.103.010.725
   Max 1st sample MIU time among all SL packets: 2021-09-02T20:37:29.986.667.725
   Min 1st sample XO time among all SL packets: 11.103011965
   Max 1st sample XO time among all SL packets: 47.986671381
   TM values from -274 to 247 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-02T20:36:53.103.010.725 (XO: 11.103011965 - 11.103317966, t=37.2 degC)  Rank: 00081
   TM values: Max =   247  Min =  -257  Ave =    -4.43  Std =    27.00
   Min TM value:-257 at an offset of 93.150 usec -> 2021-09-02T20:36:53.103.103.875 (XO: 11.103105115)
   Max TM value:247 at an offset of 122.613 usec -> 2021-09-02T20:36:53.103.133.337 (XO: 11.103134578)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 122.438 usec -> 2021-09-02T20:36:53.103.133.162 (XO: 11.103134403)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 119.813 usec -> 2021-09-02T20:36:53.103.130.537 (XO: 11.103131778)

 - Block number: 2 starting at 2021-09-02T20:37:04.004.086.725 (XO: 22.004088138 - 22.004394140, t=37.2 degC)  Rank: 00080
   TM values: Max =   235  Min =  -264  Ave =    -4.44  Std =    25.34
   Min TM value:-264 at an offset of 107.225 usec -> 2021-09-02T20:37:04.004.193.950 (XO: 22.004195364)
   Max TM value:235 at an offset of 122.000 usec -> 2021-09-02T20:37:04.004.208.725 (XO: 22.004210139)
   Max dTM/dt [1/us]:2.56e+03 at an offset of 121.863 usec -> 2021-09-02T20:37:04.004.208.587 (XO: 22.004210001)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 119.075 usec -> 2021-09-02T20:37:04.004.205.800 (XO: 22.004207214)

 - Block number: 3 starting at 2021-09-02T20:37:15.845.147.725 (XO: 33.845151514 - 33.845457515, t=37.2 degC)  Rank: 00082
   TM values: Max =   241  Min =  -246  Ave =    -4.46  Std =    25.04
   Min TM value:-246 at an offset of 120.375 usec -> 2021-09-02T20:37:15.845.268.100 (XO: 33.845271889)
   Max TM value:241 at an offset of 122.925 usec -> 2021-09-02T20:37:15.845.270.650 (XO: 33.845274439)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 122.750 usec -> 2021-09-02T20:37:15.845.270.475 (XO: 33.845274264)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 120.163 usec -> 2021-09-02T20:37:15.845.267.887 (XO: 33.845271677)

 - Block number: 4 starting at 2021-09-02T20:37:29.986.378.725 (XO: 47.986382380 - 47.986688381, t=37.2 degC)  Rank: 00082
   TM values: Max =   234  Min =  -274  Ave =    -4.57  Std =    22.67
   Min TM value:-274 at an offset of 108.988 usec -> 2021-09-02T20:37:29.986.487.712 (XO: 47.986491368)
   Max TM value:234 at an offset of 111.513 usec -> 2021-09-02T20:37:29.986.490.237 (XO: 47.986493893)
   Max dTM/dt [1/us]:2.92e+03 at an offset of 111.338 usec -> 2021-09-02T20:37:29.986.490.062 (XO: 47.986493718)
   Min dTM/dt [1/us]:-2.71e+03 at an offset of 108.800 usec -> 2021-09-02T20:37:29.986.487.525 (XO: 47.986491180)
