-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 31 22:49:06 2021 UTC
-- File 20210902_052612_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.891406 to 41.940905
-each separately from 4s to 41s and from 583050us to 940905us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.691355 in 1 packets: 18
      14.357277 in 1 packets: 36
MIU packet time from 2021-09-02T05:26:19.891.406.000 to 2021-09-02T05:26:56.940.905.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.691355 in 1 packets: 18
      14.357277 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9477087 to 42.9475861
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999238 in 1 packets: 18
      13.9999536 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210902_052612_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8391151 to 6.8394041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6912790  -- OK for SL --
   MIU time from 2021-09-02T05:26:19.891.406.000 to 2021-09-02T05:26:19.891.695.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.691355

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5306831 to 29.5309721
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3572310  -- OK for SL --
   MIU time from 2021-09-02T05:26:42.583.050.000 to 2021-09-02T05:26:42.583.339.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.357277

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.8882031 to 43.8884921
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-02T05:26:56.940.616.000 to 2021-09-02T05:26:56.940.905.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.839115100 corrected by 40bit 10MHz counter of 5.947708700 obtained at 1Hz counter of 4. Result: 4.891409342
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-02T05:26:19.891.265.725
   Max 1st sample MIU time among all SL packets: 2021-09-02T05:26:56.940.764.725
   Min 1st sample XO time among all SL packets: 4.891269066
   Max 1st sample XO time among all SL packets: 41.940768830
   TM values from -276 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-02T05:26:19.891.265.725 (XO: 4.891269066 - 4.891575067, t=37.2 degC)  Rank: 00078
   TM values: Max =   236  Min =  -224  Ave =    -4.48  Std =    20.18
   Min TM value:-224 at an offset of 116.225 usec -> 2021-09-02T05:26:19.891.381.950 (XO: 4.891385292)
   Max TM value:236 at an offset of 130.813 usec -> 2021-09-02T05:26:19.891.396.537 (XO: 4.891399879)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 130.688 usec -> 2021-09-02T05:26:19.891.396.412 (XO: 4.891399754)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 116.063 usec -> 2021-09-02T05:26:19.891.381.787 (XO: 4.891385129)

 - Block number: 2 starting at 2021-09-02T05:26:42.582.909.725 (XO: 27.582912249 - 27.583218250, t=37.2 degC)  Rank: 00082
   TM values: Max =   224  Min =  -265  Ave =    -4.56  Std =    25.21
   Min TM value:-265 at an offset of 132.425 usec -> 2021-09-02T05:26:42.583.042.150 (XO: 27.583044674)
   Max TM value:224 at an offset of 132.175 usec -> 2021-09-02T05:26:42.583.041.900 (XO: 27.583044424)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 134.838 usec -> 2021-09-02T05:26:42.583.044.562 (XO: 27.583047087)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 132.288 usec -> 2021-09-02T05:26:42.583.042.012 (XO: 27.583044537)

 - Block number: 3 starting at 2021-09-02T05:26:56.940.475.725 (XO: 41.940479829 - 41.940785830, t=37.2 degC)  Rank: 00081
   TM values: Max =   204  Min =  -276  Ave =    -4.47  Std =    19.82
   Min TM value:-276 at an offset of 132.400 usec -> 2021-09-02T05:26:56.940.608.125 (XO: 41.940612229)
   Max TM value:204 at an offset of 130.113 usec -> 2021-09-02T05:26:56.940.605.837 (XO: 41.940609942)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 134.813 usec -> 2021-09-02T05:26:56.940.610.537 (XO: 41.940614642)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 132.238 usec -> 2021-09-02T05:26:56.940.607.962 (XO: 41.940612067)
