-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Dec 28 16:02:22 2021 UTC
-- File 20210829_092827_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.441287 to 29.684776
-each separately from 5s to 29s and from 441287us to 684776us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.242911 in 1 packets: 18
MIU packet time from 2021-08-29T09:28:35.441.287.000 to 2021-08-29T09:28:59.684.776.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.242911 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8732159 to 30.8731359
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999200 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210829_092827_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3145034 to 7.3147924
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.2428310  -- OK for SL --
   MIU time from 2021-08-29T09:28:35.441.287.000 to 2021-08-29T09:28:35.441.576.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.242911

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5576234 to 31.5579124
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-29T09:28:59.684.487.000 to 2021-08-29T09:28:59.684.776.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.314503400 corrected by 40bit 10MHz counter of 6.873215900 obtained at 1Hz counter of 5. Result: 5.441289000
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-29T09:28:35.441.146.725
   Max 1st sample MIU time among all SL packets: 2021-08-29T09:28:59.684.635.725
   Min 1st sample XO time among all SL packets: 5.441148725
   Max 1st sample XO time among all SL packets: 29.684638553
   TM values from -281 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-29T09:28:35.441.146.725 (XO: 5.441148725 - 5.441454726, t=37.2 degC)  Rank: 00083
   TM values: Max =   232  Min =  -247  Ave =    -4.59  Std =    18.04
   Min TM value:-247 at an offset of 122.300 usec -> 2021-08-29T09:28:35.441.269.025 (XO: 5.441271025)
   Max TM value:232 at an offset of 124.888 usec -> 2021-08-29T09:28:35.441.271.612 (XO: 5.441273613)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 124.700 usec -> 2021-08-29T09:28:35.441.271.425 (XO: 5.441273425)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 122.125 usec -> 2021-08-29T09:28:35.441.268.850 (XO: 5.441270850)

 - Block number: 2 starting at 2021-08-29T09:28:59.684.346.725 (XO: 29.684349552 - 29.684655553, t=37.2 degC)  Rank: 00083
   TM values: Max =   236  Min =  -281  Ave =    -4.55  Std =    29.48
   Min TM value:-281 at an offset of 152.638 usec -> 2021-08-29T09:28:59.684.499.363 (XO: 29.684502190)
   Max TM value:236 at an offset of 111.188 usec -> 2021-08-29T09:28:59.684.457.912 (XO: 29.684460740)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 113.825 usec -> 2021-08-29T09:28:59.684.460.550 (XO: 29.684463377)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 111.300 usec -> 2021-08-29T09:28:59.684.458.025 (XO: 29.684460852)
