-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 01:36:01 2021 UTC
-- File 20210827_151104_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.387616 to 20.868498
-each separately from 10s to 20s and from 387616us to 868498us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.480304 in 1 packets: 18
MIU packet time from 2021-08-27T15:11:17.387.616.000 to 2021-08-27T15:11:27.868.498.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.480304 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9165091 to 21.9164760
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999669 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_151104_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3041257 to 12.3044147
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4802710  -- OK for SL --
   MIU time from 2021-08-27T15:11:17.387.616.000 to 2021-08-27T15:11:17.387.905.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.480304

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7846857 to 22.7849747
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T15:11:27.868.209.000 to 2021-08-27T15:11:27.868.498.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.304125700 corrected by 40bit 10MHz counter of 11.916509100 obtained at 1Hz counter of 10. Result: 10.387617879
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T15:11:17.387.475.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T15:11:27.868.357.725
   Min 1st sample XO time among all SL packets: 10.387477604
   Max 1st sample XO time among all SL packets: 20.868361291
   TM values from -276 to 237 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T15:11:17.387.475.725 (XO: 10.387477604 - 10.387783605, t=37.2 degC)  Rank: 00081
   TM values: Max =   237  Min =  -276  Ave =    -4.58  Std =    28.57
   Min TM value:-276 at an offset of 122.375 usec -> 2021-08-27T15:11:17.387.598.100 (XO: 10.387599979)
   Max TM value:237 at an offset of 120.100 usec -> 2021-08-27T15:11:17.387.595.825 (XO: 10.387597704)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 124.675 usec -> 2021-08-27T15:11:17.387.600.400 (XO: 10.387602279)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 122.150 usec -> 2021-08-27T15:11:17.387.597.875 (XO: 10.387599754)

 - Block number: 2 starting at 2021-08-27T15:11:27.868.068.725 (XO: 20.868072290 - 20.868378291, t=37.2 degC)  Rank: 00091
   TM values: Max =   218  Min =  -272  Ave =    -4.44  Std =    22.90
   Min TM value:-272 at an offset of 120.888 usec -> 2021-08-27T15:11:27.868.189.612 (XO: 20.868193178)
   Max TM value:218 at an offset of 120.613 usec -> 2021-08-27T15:11:27.868.189.337 (XO: 20.868192903)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 123.250 usec -> 2021-08-27T15:11:27.868.191.975 (XO: 20.868195540)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 120.738 usec -> 2021-08-27T15:11:27.868.189.462 (XO: 20.868193028)
