-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 27 00:03:19 2021 UTC
-- File 20210827_134841_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.509377 to 15.508943
-each separately from 11s to 15s and from 508654us to 509666us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.998988 in 1 packets: 18
MIU packet time from 2021-08-27T13:48:55.509.377.000 to 2021-08-27T13:48:59.508.943.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.998988 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8671347 to 16.8671215
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999868 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_134841_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3765125 to 13.3768015
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.9989750  -- OK for SL --
   MIU time from 2021-08-27T13:48:55.509.377.000 to 2021-08-27T13:48:55.509.666.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.998988

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3757765 to 17.3760655
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T13:48:59.508.654.000 to 2021-08-27T13:48:59.508.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.376512500 corrected by 40bit 10MHz counter of 12.867134700 obtained at 1Hz counter of 11. Result: 11.509379481
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T13:48:55.509.236.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T13:48:59.508.802.725
   Min 1st sample XO time among all SL packets: 11.509239205
   Max 1st sample XO time among all SL packets: 15.508805455
   TM values from -263 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T13:48:55.509.236.725 (XO: 11.509239205 - 11.509545206, t=37.2 degC)  Rank: 00082
   TM values: Max =   204  Min =  -263  Ave =    -4.51  Std =    18.34
   Min TM value:-263 at an offset of 128.725 usec -> 2021-08-27T13:48:55.509.365.450 (XO: 11.509367931)
   Max TM value:204 at an offset of 128.450 usec -> 2021-08-27T13:48:55.509.365.175 (XO: 11.509367656)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 131.088 usec -> 2021-08-27T13:48:55.509.367.812 (XO: 11.509370293)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 128.550 usec -> 2021-08-27T13:48:55.509.365.275 (XO: 11.509367756)

 - Block number: 2 starting at 2021-08-27T13:48:59.508.513.725 (XO: 15.508516454 - 15.508822455, t=37.2 degC)  Rank: 00081
   TM values: Max =   211  Min =  -257  Ave =    -4.46  Std =    26.19
   Min TM value:-257 at an offset of 173.038 usec -> 2021-08-27T13:48:59.508.686.763 (XO: 15.508689492)
   Max TM value:211 at an offset of 131.663 usec -> 2021-08-27T13:48:59.508.645.387 (XO: 15.508648117)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 131.500 usec -> 2021-08-27T13:48:59.508.645.225 (XO: 15.508647954)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 128.938 usec -> 2021-08-27T13:48:59.508.642.662 (XO: 15.508645392)
